Cellular transfer and AFM imaging of cancer cells using Bioimprint
© Muys et al; licensee BioMed Central Ltd. 2006
Received: 11 August 2005
Accepted: 22 January 2006
Published: 22 January 2006
A technique for permanently capturing a replica impression of biological cells has been developed to facilitate analysis using nanometer resolution imaging tools, namely the atomic force microscope (AFM). The method, termed Bioimprint™, creates a permanent cell 'footprint' in a non-biohazardous Poly (dimethylsiloxane) (PDMS) polymer composite. The transfer of nanometer scale biological information is presented as an alternative imaging technique at a resolution beyond that of optical microscopy. By transferring cell topology into a rigid medium more suited for AFM imaging, many of the limitations associated with scanning of biological specimens can be overcome. Potential for this technique is demonstrated by analyzing Bioimprint™ replicas created from human endometrial cancer cells. The high resolution transfer of this process is further detailed by imaging membrane morphological structures consistent with exocytosis. The integration of soft lithography to replicate biological materials presents an enhanced method for the study of biological systems at the nanoscale.
Currently, optical microscopy techniques are the primary method for cell surface visualization, with microscopic characteristics of cells traditionally used for diagnosis and classification of cancers . However, because the differences in characteristics can be subtle, accurate detection can be challenging and ambiguous . A drawback of analysis using focused light microscopy is the fundamental diffraction limit, which at its optimum imposes an attainable spatial limit of 180 nm in the focal plane and 500 nm along the optical axis [3, 4].
Imaging tools, such as the atomic force (AFM) and scanning electron (SEM) microscopes, are investigated for their ability to provide topographical information at a resolution far superior to optical methods [5, 6]. Despite having the potential to image numerous diseases, cancers and pathogens, nanoscale analytical tools have not been efficiently utilized in mainstream biological research. Difficulties associated with imaging soft, living biological material in situ is challenging and remains a delicate and time consuming task, which while effective, is inefficient for the analysis and evaluation of large cell populations . Predictions on the attainable resolution when imaging cell surfaces by AFM in liquid, regardless of living or fixed, is generally considered in be in the order of 50–500 nm [8–10]. Resolution limiting factors  include cell-to-substrate attachment, cell type, topographic complexity, surface composition and tip indentation into the soft biological material. In a probing-based imaging environment soft biomaterials are susceptible to structural movement and deformation caused by intermittent contact by the sharp AFM tip . The tip apex is a crucial resolution limiting factor in AFM investigations and previous biological analysis has been limited to blunter probes for fear of membrane penetration .
Time consuming preparation procedures used for air and vacuum imaging environments require dehydration and fixation, which can also cause deformation and artifacts . Although fluid-based AFM or SEM imaging attempts to address these issues by maintaining physiological conditions, factors such as scanning time, probe or electron interaction, and dampening effects are difficulties limiting these useful techniques. Consequently, nano-imaging as an analytical tool in biology remains under-utilized.
In the semiconductor industry, lithography enables the high resolution pattern transfer for the fabrication of nanoscale structures and devices. Recently, nanoimprint, a form of soft lithography, has been added as a candidate for next generation lithography; a successor to photolithography for pattern replication in the manufacturing of integrated circuits [15, 16]. Soft lithography functions by contacting a structured template into a soft liquified polymer material, enabling a permanent replica to be fabricated after curing.
By using a technique, termed Bioimprint™, biological cells are directly integrated with soft lithography fabrication processes to create cell impressions in a robust storage medium for subsequent analysis using nano-imaging tools. In the process, a biocompatible liquid polymer is brought into contact with a cell before curing to create a negative replica.
This paper presents an alternative method for studying biological cells using a Bioimprint™ technique with AFM analysis, to enable the high-resolution replication and imaging of the surface topography of human endometrial cancer cells. As non-malignant endometrial cells were not immediately available as controls this work represents a preliminary study.
Exocytosis has been described in numerous ways with wide speculation governing both the underlying mechanisms driving membrane fusion as well as membrane topology. It is accepted that fusion begins by a granule or vesicle from within the cell docking at the membrane to release its contents. The manner by which the contents of the granule are released remains debated, and there are arguments supporting both total and transient fusion as described by 'fuse-and-collapse' and 'kiss-and-run' mechanisms, respectively. Visual verification has however been limited, partly due to the difficulties with imaging living or structurally intact cells at high resolution, and the lack of well-defined protocols and methods integrating nano-imaging tools with biology. The dimple model [17–21] predicts that exocytosis is initiated by membrane fusion; in which, a scaffold built into the membrane dilates to create a dimple site, where subsequently an underlying granule docks to create a fusion pore and release its contents. While other models [22, 23] for membrane fusion exist this remains one of the most convincing and well documented models for exocytosis.
An additional feature depicted in Fig. 1(a) is the outline of a spherical form impacting on the cell membrane, which is assumed to be the nucleus (N). In Fig. 1(b) a scope trace reveals the impact of the underlying nucleus on the membrane, causing a distorted effect indicated by points (1),(2). Weyn et al.  have investigated the dehydration effects on malignant mesothelioma cells by AFM and reported a much harder and uniform indentation profile over the entire cell, whereas hydrated cells have a more rounded and smooth surface. Nuclei collapse was also a possible feature resulting from dehydration effects and though some cells demonstrated nuclei submersion, this did not occur in every cell imprinted.
Much is yet to be known about the nature of endometrial cancer cells and until now there has not been a reliable, simple method for visualizing cell topography in air, at high resolution without fixation and dehydration. Being able to directly view membrane structures regulated by exocytosis will enable researchers to analyze the secretory nature and response of cells, yielding insights into drug responses and effects. Considerable variability in the sizes of dimple depressions and ruptures, as well as dynamic formation and grouping of these structures around the nucleus, illustrates that cells have diverse morphologies.
There was an inconsistency in the degree of deformities in the cancer cells: Reasons for the varying nuclei forms seen could potentially be explained by the weighted force of the polymer, acting to press the membrane down and casting the nucleus as a structure protruding from the cell. However, the ability to view and potentially characterize the effects of cancer mutations at both micron and nanoscales presents a remarked improvement over conventional optics. In addition to the high resolution imaging enabled by the AFM, the ability to image accurately in 3-dimensions presents a significant advantage over radiation scattered imaging devices. The sensitivity of the AFM and precise transfer of cell topography into a polymer provides a method with the ability to overcome the current difficulties of imaging biological materials by the AFM.
Advantages of Bioimprint™ extend beyond simple lithographic process replication, with benefits such as prolonged storage and analytical adaptability without lose of resolution. Additionally, being a non-biohazardous substance, impressions of pathogenic material, infected cells and other biological samples can be transported or exchanged for analysis without contamination concerns. This would reduce the need for complicated and lengthy documental approval, which is required by governments and institutions, and would facilitate out-of-house analysis. Further gain is the ability to keep patient records using an indirect specimen, without the need for expensive storage or contamination equipment.
Usefulness of Bioimprint™ can especially be realized when used as a complementary technique with conventional optical imaging, and as alternative method for those requiring strict sample preparation, such as chemical fixation and dehydration in order to visualize cell topology by AFM  or SEM . Such applications could employ immunohistochemistry methodology on the actual cell, prior to replicating the cell topography using Bioimprint™. This combined physical and chemical approach may yield a better understanding into cell functionality and mechanics.
Artifacts in the form of bubbles caught trapped between the cell and polymer and ripping of the cell membrane were also observed and readily identifiable. A crucial factor in imprinting the cell structure is the amount of fluid remaining on the surface prior to polymeric application. Absence of a thin layer of medium above the cells inevitably causes aridity, and resulted in many Bioimprint™ replicas displaying characteristic nucleus dehydration artifacts. On the other hand, too much fluid will create an interfacial layer that impedes the transfer of high resolution features.
A limitation in the current methodology used to fabricate imprints is that the process lacks the control required for fabricating accurate and consistent cell replicas. Imprinting conditions are too slow and a large proportion of the cell population failed to be accurately replicated due to significant dehydration effects. Using the heat curing PDMS polymer, the replicas are in fact a 'time-averaged' rather than a 'single-shot' capture impression, in which the polymer conforms to the cell structure. Artifacts are inevitably being introduced by the cellular response to the polymer and curing conditions, which are most noticeably shown by the affects on the nucleus.
Currently, efforts are being concentrated around developing a rapidly U.V. curable polymer formed as a thin pre-spun layer, which is imprinted rather than poured above the cells in a bulk . While the U.V. light is undoubtedly detrimental to cell physiology, the time taken to replicate is 100-fold shorter than the conventional PDMS composite, enabling a more accurate representation of the living cell to be replicated. Preliminary results show a reduction in the number of cells displaying nuclei submersion or dehydrating effects and an improved resolution transfer.
A soft lithographic technique for creating replica cell impressions with nanoscale information transfer has been introduced and tested on human endometrial cancer cells. By creating a cell 'footprint' in a stable solidified polymer, a permanent non-biohazardous record can be kept and analyzed at high resolution using the AFM. Bioimprint™ overcomes many of the inherent difficulties associated with cellular imaging by AFM and advances their integration as investigative tools in biology. With visual verification ultimately being the mainstay for cancer diagnosis, a method facilitating the use of imaging at potentially atomic resolution could be used more to characterize morphological abnormalities at the nanoscale. Though at this time, it is difficult to deduce if the varying shapes and forms are potential characteristics linked to malignant cell mutations, or if they are artifacts induced from the Bioimprint™ replication process. This study reports preliminary work in the areas of cellular replication and endometrial cancer cell imaging by atomic force microscopy.
Human endometrial cancer cells were cultured in accordance with institutional guidelines of the Christchurch School of Medicine and Health Sciences, University of Otago, New Zealand, after ethical approval and appropriate informed consent. The preparation of cells were as follows: Endometrial adenocarcinoma tissues were harvested from women undergoing hysterectomy, and non-myometrial biopsies were taken from the opened uterus tumor area. Tissues were then digested in collagenase-A (1 mg/ml), and the cells dispersed, and cultured overnight in medium consisting of alpha-MEM containing 1 % penicillin/streptomycin, 0.1 % BSA and 10 % fetal calf serum.
The hardened Bioimprint™ impressions were analyzed by an AFM (DI 3100, Veeco Instruments, Santa Barbara, CA) in tapping mode using triangular non-contact cantilevers (NSC11, MikroMasch, Estonia), which were typically operated between 0.6–1 Hz at a resonant frequency of ~315 kHz with a nominal sub-10 nm radius of curvature and a force constant 48 N/m. To recover the original cell orientation, positive replicas are made by digitally inverting the AFM scans of the impressions/molds/negative replica, which were made by the cells when imprinted in the polymer.
Authors wish to thank all women from the Christchurch Women's Hospital, whom without their donations this research would not of been possible. We further acknowledge the significant contributions made by nurse Dianne Harker and her efforts made ensuring the ethical processing of all tissues. This research was supported by the MacDiarmid Institute for Advanced Materials and Nanotechnology.
- Papanicolaou GN, Traut HF: Am J Obstet Gynecol. 1941, 42: 193-206.Google Scholar
- Hamby L: Gene expression patterns and breast cancer. Cancer Genetics News, Spring. 2002, 4: 1.Google Scholar
- Hell SW: Toward fluorescence nanoscopy. Nat Biotech. 2003, 21: 1347-1355. 10.1038/nbt895.View ArticleGoogle Scholar
- Pawley JB: Handbook of Biological Confocal Microscopy. New York, Plenum Press; 1995.View ArticleGoogle Scholar
- Binnig G, Quate CF, Gerber Ch: Atomic force microscope. Phys Rev Let. 1986, 56: 930-933. 10.1103/PhysRevLett.56.930.View ArticleGoogle Scholar
- Binnig G, Gerber Ch, Stoll E, Albrecht TR, Quate CF: Atomic resolution with atomic force microscope. Europhys Lett. 1987, 3: 1281.View ArticleGoogle Scholar
- Dufrêne YF: Recent progress in the application of atomic force microscopy imaging and force spectroscopy to microbiology. Curr Opin Microbiol. 2003, 6: 317-323. 10.1016/S1369-5274(03)00058-4.View ArticleGoogle Scholar
- Fritz M, Radmacher M, Gaub HE: Granula motion and membrane spreading during activation of human platelets imaged by atomic force microscopy. Biophys J. 1994, 66: 1328-1334.View ArticleGoogle Scholar
- Radmacher M: Measuring the elastic properties of biological samples with the AFM. IEE Eng Med Biol. 1997, 47-57. 10.1109/51.582176.Google Scholar
- Shao Z, Yang J, Somlyo AP: Biological atomic force microscopy: from microns to nanometers and beyond. Annu Rev Cell Dev Biol. 1995, 11: 241-65. 10.1146/annurev.cb.11.110195.001325.View ArticleGoogle Scholar
- Le Grimellec C, Lesniewska E, Giocondi MC, Finot E, Vie V, Goudonnet JP: Imaging of the surface of living cells by low-force contact-mode atomic force microscopy. Biophys J. 1998, 75 (2): 695-703.View ArticleGoogle Scholar
- Häberle W, Hörber JKH, Binnig G: Force microscopy on living cells. J Vac Sci Technol B. 1990, 9 (2): 1210-1213. 10.1116/1.585206.View ArticleGoogle Scholar
- You HX, Yu L: Atomic force microscopy imaging of living cells: progress, problems and prospects. Methods Cell Sci. 1999, 21 (1): 1-17. 10.1023/A:1009876320336.View ArticleGoogle Scholar
- Garg A, Kokkoli E: Characterizing partilculate drug-deliver carriers with atomic force microscopy. IEEE Engineering in Biology and Medicine Mag. 2005, 87-95. 10.1109/MEMB.2005.1384106.Google Scholar
- Zhao XM, Xia Y, Whitesides GM: Fabrication of three-dimensional micro-structures: Micro-transfer molding. Adv Mater. 1996, 8: 420-424. 10.1002/adma.19960080511.View ArticleGoogle Scholar
- Chou SY, Krauss PR, Renstrom PJ: Imprint lithography with 25-nanometer resolution. Science. 1996, 272: 85-87.View ArticleGoogle Scholar
- Ornberg RL, Reese TS: Beginning of exocytosis captured by rapid-freezing of Limulus ame-bocytes. J Cell Biol. 1981, 90: 40-10.1083/jcb.90.1.40.View ArticleGoogle Scholar
- Nanavati C, Markin VS, Oberhauser A, Fernandez JM: The exocytotic fusion pore as a protein-supported lipidic structure. Biophys J. 1992, 63: 1118.View ArticleGoogle Scholar
- Fernandez JM: Cellular and molecular mechanics by atomic force microscopy: Capturing the exocytotic fusion pore in vivo?. Proc Natl Acad Sci USA. 1997, 94: 9-10.1073/pnas.94.1.9.View ArticleGoogle Scholar
- Chandler DE: Membrane fusion as seen in rapidly frozen secretory cells. Ann NY Acad Sci. 1992, 635: 234.View ArticleGoogle Scholar
- Chandler DE, Heuser JE: Arrest of membrane fusion events in mast cells by quick freezing. J Cell Biol. 1980, 86: 666-10.1083/jcb.86.2.666.View ArticleGoogle Scholar
- Cho SJ, Jeftinija K, Glavaski A, Jeftinija S, Jena BP, Anderson LL: Structure and dynamics of the fusion pores in live GH-secreting cells revealed using atomic force microscopy. Endocrin. 1997, 143: 1144-1148. 10.1210/en.143.3.1144.View ArticleGoogle Scholar
- Jena BH, Cho S-J, Jeremic A, Stromer MH, Abu-Hamdah R: Structure and Composition of the Fusion Pore. Biophys J. 2003, 84 (2): 1337-1343.View ArticleGoogle Scholar
- Weyn B, Kalle W, Kumar-Singh S, Van Marck E, Tanke H, Jacob W: Atomic force microscopy: influence of air drying and fixation on the morphology and viscoelasticity of cultured cells. J Microscopy. 1998, 189 (2): 172-180. 10.1046/j.1365-2818.1998.00299.x. (9)View ArticleGoogle Scholar
- Zink D, Fischer AH, Nickerson JA: Nature Reviews. Nucl Struct Canc Cell. 2004, 4: 677-687.Google Scholar
- Ishioka S-i, Sagae S, Ito E, Kudo R: Ultrastructural study of benign, low-malignant potential (LMP), and malignant ovarian tumors. Medical Electron Microscopy. 2004, 37 (1): 37-44. 10.1007/s00795-003-0189-0.View ArticleGoogle Scholar
- El Kirat K, Burton I, Dupres V, Dufrene YF: Sample preparation procedures for biological atomic force microscopy. J Microscopy. 2005, 218 (3): 199-207. 10.1111/j.1365-2818.2005.01480.x.View ArticleGoogle Scholar
- Hayat MA: Principles and Techniques of Electron Microscopy: Biological Applications: Ch.1–3,6,7. 2000, Cambridge University Press, UK, 4Google Scholar
- Muys JJ, Alkaisi MA, Evans JJ: Bioimprint: Nanoscale analysis by replication of cellular topography using soft lithography. J Biomed Nano In press.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.