- Open Access
Non-destructive on-chip cell sorting system with real-time microscopic image processing
© Takahashi et al; licensee BioMed Central Ltd. 2004
- Received: 13 December 2003
- Accepted: 03 June 2004
- Published: 03 June 2004
Studying cell functions for cellomics studies often requires the use of purified individual cells from mixtures of various kinds of cells. We have developed a new non-destructive on-chip cell sorting system for single cell based cultivation, by exploiting the advantage of microfluidics and electrostatic force. The system consists of the following two parts: a cell sorting chip made of poly-dimethylsiloxane (PDMS) on a 0.2-mm-thick glass slide, and an image analysis system with a phase-contrast/fluorescence microscope. The unique features of our system include (i) identification of a target from sample cells is achieved by comparison of the 0.2-μm-resolution phase-contrast and fluorescence images of cells in the microchannel every 1/30 s; (ii) non-destructive sorting of target cells in a laminar flow by application of electrostatic repulsion force for removing unrequited cells from the one laminar flow to the other; (iii) the use of agar gel for electrodes in order to minimize the effect on cells by electrochemical reactions of electrodes, and (iv) pre-filter, which was fabricated within the channel for removal of dust contained in a sample solution from tissue extracts. The sorting chip is capable of continuous operation and we have purified more than ten thousand cells for cultivation without damaging them. Our design has proved to be very efficient and suitable for the routine use in cell purification experiments.
- Cell Sorting
- Electrostatic Force
- Microfluidic Channel
Developing of a cell based screening assay often requires identification and isolation of particular cells from a mixture of various kinds of cells. Moreover, in order to obtain reproducible data on cells, reliable and non-destructive purification of cells is essential. Efficient and rapid sorting of cells has been accomplished with techniques such as fluorescence-activated cell sorting (FACS) , magnetic-activated cell separation (MACS), automated single-cell sorting using dual-beam optical trapping, differential adhesion cell sorting , and micro-fabricated fluorescence-activated cell sorting . These conventional cell sorters can be used for purification of individual cells, but are not ideal techniques and have a number of disadvantages.. For example, FACS can damage cells during destructive droplet generation, and the detection (based on the non-direct scattering) has poor cell recognition performance. Other conventional techniques also have disadvantages with regard to their cost, efficiency, response speed, separate resolution, and adaptability. Furthermore, if cells are to be cultivated after subsequent sorting, the damage to cells caused by to the sorting process should be minimized. Advantages offered by application of microfluidics include reduced sample volumes and reaction times, lowered space requirements and operational costs . In micro- and nano-technology liquid flow is mostly laminar and this further facilitates making microscopic sorting devises for biomedical applications. We have developed new on-chip microfluidic cell sorting system, which overcomes problems associated with the conventional cell sorting techniques. Our system directly monitors and recognizes cells with specific index using phase-contrast/fluorescence microscopy and image processing, and it can rapidly and safely sort them according to the index (shape, and spatial distribution of fluorescent dye of the specific antibody marker, and so on) in a laminar flow using the electrostatic force. The electric force is applied only to the cells which have to be removed to waste; the target cells simply flow down the laminar flow and do not receive any stimulation from non-contact forces. In this paper, we explain the design of the on-chip cell sorter system and give the results of an experimental implementation of our design.
Design of on-chip cell sorting system
Cell sorter chip design and process
Principle of sorting procedure
Our cell sorting system differs form other available systems in that only the cells, which have to be removed, were affected by the electrostatic force. The target cells had no force applied and therefore had minimal or no damage done to them. Moreover, the area of the electric field is limited to the cell sorting area, because the electric current flows only between the two gel electrodes. Thus cells out of the area receive no significant damage from the electric stimulation.
Installation of phase-contrast image recognition into the sorting system
Composition of agar-gel for low resistance electrode
Construction of filter for purification of cells and removal of dust in the sample
Sorting accuracy of the on-chip cell sorter system
We have developed anon-chip cell sorting system that can separate target cells mildly and reliably using our original method and bio-compatible materials, such as agarose gel that is used as the electrode for the generation of electrostatic force in the flow channel. Cell sorting is based on the image analysis processed automatically by computer. A micro-sized filter for a dust-free flow is fabricated in order to purify sample cells and avoid blocking of the channels by dust particles. Using our system we have achieved 100% accurate sorting. We plan to extend our approach to single cell analysis and regenerative medicine.
Financial support, in part by the Japan Science and Technology Organization (JST) and by Grants-in-Aids for Science Research from the Ministry of Education, Science and Culture of Japan, is gratefully acknowledged.
- Bonner WA, Hulett HR, Sweet RG: Fluorescence activated cell sorting. Rev Sci Instrument. 1972, 43: 404-409. 10.1063/1.1685647.View ArticleGoogle Scholar
- Steinberg M: Reconstruction of tissues by dissociated cells. Science. 1963, 141: 401-408.View ArticleGoogle Scholar
- Fu AY: A microfabricated fluorescence-activated cell sorter. Nature Biotech. 1999, 17: 1109-1111. 10.1038/15095.View ArticleGoogle Scholar
- Barry R, Ivanov D: Microfluidics in Biotechnology. J Nanobiotechnology. 2004, 2: 2-10.1186/1477-3155-2-2.View ArticleGoogle Scholar
- Qin D, Xia Y, Whitesides GM: A rapid prototyping method for generating patterns and structures with feature sizes larger than 20 μm. Adv Mater. 1996, 8: 917-919.View ArticleGoogle Scholar
- Duffy DC, McDonald JC, Schueller OJA, Whitesides GM: Rapid prototyping microfluidics systems in poly(dimetylsiloxane). Anal Chem. 1998, 70: 4974-4984. 10.1021/ac980656z.View ArticleGoogle Scholar
- Xia Y, Whitesides GM: Soft lithography. Angew Chem Int Ed Engl. 1998, 37: 550-575. 10.1002/(SICI)1521-3773(19980316)37:5<550::AID-ANIE550>3.3.CO;2-7.View ArticleGoogle Scholar
- Jackman RJ, Duffy DC, Ostuni E, Willmore ND, Whitesides GM: Fabricating large arrays of microwells with arbitrary dimensions and filling them using discontinuous dewetting. Anal Chem. 1998, 70: 2280-2287. 10.1021/ac971295a.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article: verbatim copying and redistribution of this article are permitted in all media for any purpose, provided this notice is preserved along with the article's original URL.