- Open Access
Detection and quantification of bacterial biofilms combining high-frequency acoustic microscopy and targeted lipid microparticles
Journal of Nanobiotechnology volume 12, Article number: 24 (2014)
Immuno-compromised patients such as those undergoing cancer chemotherapy are susceptible to bacterial infections leading to biofilm matrix formation. This surrounding biofilm matrix acts as a diffusion barrier that binds up antibiotics and antibodies, promoting resistance to treatment. Developing non-invasive imaging methods that detect biofilm matrix in the clinic are needed. The use of ultrasound in conjunction with targeted ultrasound contrast agents (UCAs) may provide detection of early stage biofilm matrix formation and facilitate optimal treatment.
Ligand-targeted UCAs were investigated as a novel method for pre-clinical non-invasive molecular imaging of early and late stage biofilms. These agents were used to target, image and detect Staphylococcus aureus biofilm matrix in vitro. Binding efficacy was assessed on biofilm matrices with respect to their increasing biomass ranging from 3.126 × 103 ± 427 UCAs per mm2 of biofilm surface area within 12 h to 21.985 × 103 ± 855 per mm2 of biofilm matrix surface area at 96 h. High-frequency acoustic microscopy was used to ultrasonically detect targeted UCAs bound to a biofilm matrix and to assess biofilm matrix mechanoelastic physical properties. Acoustic impedance data demonstrated that biofilm matrices exhibit impedance values (1.9 MRayl) close to human tissue (1.35 - 1.85 MRayl for soft tissues). Moreover, the acoustic signature of mature biofilm matrices were evaluated in terms of integrated backscatter (0.0278 - 0.0848 mm-1 × sr-1) and acoustic attenuation (3.9 Np/mm for bound UCAs; 6.58 Np/mm for biofilm alone).
Early diagnosis of biofilm matrix formation is a challenge in treating cancer patients with infection-associated biofilms. We report for the first time a combined optical and acoustic evaluation of infectious biofilm matrices. We demonstrate that acoustic impedance of biofilms is similar to the impedance of human tissues, making in vivo imaging and detection of biofilm matrices difficult. The combination of ultrasound and targeted UCAs can be used to enhance biofilm imaging and early detection. Our findings suggest that the combination of targeted UCAs and ultrasound is a novel molecular imaging technique for the detection of biofilms. We show that high-frequency acoustic microscopy provides sufficient spatial resolution for quantification of biofilm mechanoelastic properties.
Bacterial biofilms are three-dimensional extracellular matrices composed of carbohydrates, proteins and exopolysaccharides [1–6] that develop on solid–liquid or solid-air interfaces in the body [3, 7]. Biofilms consist of bacterial cells and matrix proteins. The majority of biofilms contain 10% or less of bacterial cells and over 90% matrix . Biofilm matrices are highly conserved dynamic structures. Initiation of a biofilm matrix occurs by a transient interaction of bacteria with a surface followed by an adhesive stage that allows for microcolony formation and a subsequent growth and maturation stage. The complexity of biofilms allows bacteria cells to survive a multitude of environments and promotes cell dispersion to colonize new areas. These matrices may form on medical devices or fragments of dead tissue [3, 9–12]. Clinically, biofilms may occur during chemotherapy and infectious diseases such as endocarditis [6, 13–16].
Biofilm associated infections are resistant to treatment and recur even after repeated antibiotic therapy. One primary issue is that established biofilm matrices act as diffusion barriers and actively bind up antibiotics and antibodies thereby providing increased resistance. Overall killing bacteria that are surrounded by a microbial biofilm require up to 1000 times higher concentrations of antibiotics than those without a surrounding biofilm [17–20]. Thus, detecting, treating and inhibiting biofilm formation inside the body is a key medical challenge.
Moreover, in the clinical setting antibiotic therapy efficacy is decreased in the presence of an established biofilm making early detection critical. For example, infective endocarditis may occur due to chronic infection, has a poor prognosis and is associated with high mortality rates [14–16, 21]. Indeed, there are significant diagnostic challenges for endocarditis that are attributed to the inaccessibility of intra-cardiac biofilms and the non-specific nature of the clinical symptoms . Although echocardiography permits non-invasive detection of biofilms  it has significant limitations in the detection of early biofilm matrix formation. In addition, clinical diagnosis primarily occurs after biofilm matrices are fully established, thereby significantly decreasing available treatment options. Therefore, early detection is a crucial component of diagnosis; however no current diagnostic methodology is available that clearly delineates early and late stage matrices.
Ultrasound is an effective method for imaging biofilms in vitro [24–28]. One method used to enhance biofilm detection is the addition of UCAs (encapsulated gas bubbles), which provide a unique acoustic scattering signature thereby significantly enhancing imaging capabilities . Furthermore, linking a ligand to a contrast agent’s outer membrane aids in UCAs binding to tissue and is crucial in delineating disease specific regions from surrounding healthy tissue [30–35].
In this study, ligand-targeted UCAs were used as a novel method for pre-clinical non-invasive molecular imaging of early and late stage biofilms. These agents were used to target and detect Staphylococcus aureus (S. aureus) biofilm formation. Binding efficacy was assessed on established biofilms as a function of surface area. A combination of acoustic and optical microscopy was used to quantify the mechanical and structural properties of a three dimensional biofilm matrix. We show that high-frequency scanning acoustic microscopy (SAM) provides sufficient high spatial resolution for imaging and quantification of biofilm thickness and mechanoelastic properties.
Biofilm formation occurs when bacterial cells enter the body and attach to the underlying endothelium or tissues. Over time, biofilms form a protective three dimensional matrix that results in lower antibody efficacy in vivo (Figure 1). Biofilm surface areas were assessed by epifluorescence microscopy images of stained S. aureus biofilms at various time points (Figure 2A). Biofilm matrix surface area doubled during the first 12 hours after inoculation (growing from 26.85 mm2 ± 6.72 mm2 to 51.7 mm2 ± 2.12 mm2 at 12 h and 24 h respectively; p < 0.05). Similar growth patterns were observed through 96 hours (68.95 mm2 ± 4.6 mm2, 122.2 mm2 ± 8.56 mm2 and 179.2 mm2 ± 2.97 mm2 for 48 h, 72 h and 96 h respectively; p < 0.05). These data suggest that biofilm matrices are produced over time in our in vitro culture system.
To determine whether targeted UCAs bind to a biofilm matrix in vitro, we next examined whether targeted ultrasound contrast agents (UCAs) bound to the biofilm matrix over time. We observed an increase in the binding rate of targeted UCAs to the biofilm matrix (Figure 2B). We tested whether labeled targeted UCAs were detectable upon a labeled biofilm matrix. Tetramethylrhodamine isothiocyanate (TRITC)-streptavidin conjugated UCAs (red staining) were detectable from fluorescein isothiocyanate (FITC) anti-WGA labeled matrix (green staining). At the 12 h time point 1.109 × 103 ± 142 UCAs were bound to the biofilm. The number of bound bubbles significantly increased to 3.126 × 103 ± 427 over the following 12 h. Between 24 h and 72 h labeled UCAs binding increased (5.042 × 103 ± 285 UCAs at 48 h, 7.563 × 103 ± 142 at 72 h; p < 0.05). Between 72 h and 96 h a significant increase in targeted UCAs was observed (7.563 × 103 ± 142 to 21.985 ± 855 at 96 h; p < 0.05) suggesting that binding increases in correlation with biofilm matrix surface area. Fluorescence images stained for S. aureus biofilm matrix at various time points (Figure 2C) confirms that targeted UCAs bound more as the biofilm matrix increased over 96 hours.
Developing a non-invasive diagnostic method to detect biofilm matrices early (or at initial stages) would be a valuable clinical tool if the targeted agents could be detected acoustically. Because we determined that targeted UCAs bound proportionately to biofilm matrix mass we next assessed whether ultrasound could be used to detect targeted UCAs in vitro. The center frequency for the ultrasonic evaluation was 100 MHz  allowing for a rigorous quantification of biofilm matrix mechanoelastic properties in our in vitro biofilm culture system (Table 1).
For the physical evaluation of biofilm matrix properties (density, acoustic attenuation, ultrasound velocity, acoustic impedance and bulk modulus) a time-resolved high-frequency scanning acoustic microscope was used (Fraunhofer IBMT, St. Ingbert, Germany; Table 2). For imaging, an acoustic lens is triggered by a piezoelectric transducer that emits and receives highly focused sound waves and resolves them along the time axis (Figure 3A). The echoes reflected off the sample surface, the substrate and the interface between the sample and the substrate were taken into consideration for mechanoelastic quantification. The acoustic lens is mounted on top of the stage of a Zeiss Axiovert M200, inverted light microscope (Figure 3B). This custom arrangement [36, 37], where the optical microscope objective and the acoustic lens are confocally aligned allows for corresponding optical (or fluorescence) imaging and therefore facilitating novel simultaneous acoustical and optical evaluation of specimen.
S. aureus mature biofilms at day five were ultrasonically and fluorescently evaluated (Figure 4). UCAs were conjugated with TRITC-labeled streptavidin that allowed for the detection of the corresponding fluorescent signal. Because the acoustic lens is confocally aligned we were able to overlay the corresponding acoustic and fluorescent signals. In each case three or more different regions were scanned covering a total surface of 1 mm2 for each independent acquisition. For the same time point, fluorescence and optical images were acquired from the identical regions (Figure 4). The grey colored area corresponds to the acquired acoustic dataset of the biofilm matrix while the inset depicts a fluorescent image of a partial area within that ultrasonically acquired region. The epifluorescent images were comparable in terms of specimen location and provided different complementary information on the biofilm structure and mechanical properties (Table 1). Microbubbles (Targeson, San Diego, CA, USA) were 2–3 μm in diameter and were detected based on fluorescence and acoustic signals (Figure 4).
We next examined whether regions of biofilm matrices can be delineated based on targeted and bound or non-targeted, non-bound UCAs. When no biofilm mass was present the targeted UCAs remain unbound as there is no ligand for them to bind to (Figure 5A). As biofilm matrix formation progresses, targeted UCAs bind to the ligand present in the biofilm matrix (Figure 5B). Targeted UCAs bound to the biofilm matrix scattered sound and produced a detectable acoustic signature [38, 39], which correlates with a biofilm matrix. The images shown in Figure 4 depict both the corresponding optical and acoustic images of the UCAs. The acoustic image in Figure 6A depicts UCAs reflectivity in backscatter intensity and their spatial location is depicted as red signals in the fluorescent image. Furthermore, Figure 6A demonstrates that bound UCAs provide a stronger backscatter intensity as compared to the regions of biofilm matrix alone.
Based on linear acoustics  the integrated backscatter coefficient and acoustic attenuation were calculated for regions of bound and unbound UCAs. The mean integrated backscatter coefficient (IBSC; Figure 6A) was determined for frequencies ranging from 97 MHz to 104 MHz from the measured biofilm regions in which targeted UCAs were bound versus biofilm matrix alone. This frequency range corresponded to 36 points at a sampling frequency of 400 MHz while the scanning of the region of interests (ROIs) was performed with step sizes in the order of 10 μm in the x- and y-direction respectively. The acquired mean values of the IBSCs for the ROIs that remained bound to targeted UCAs range from 0.0278 mm-1 × sr-1 to 0.0848 mm-1 × sr-1 while the values for the standard deviation (StDev) vary from 0.0016 mm-1 × sr-1 to 0.0043 mm-1 × sr-1. The evaluation of the ROIs corresponding to the matrix without UCAs yielded for the IBSCs mean values in the range from 0.0167 mm-1 × sr-1 to 0.0694 mm-1 × sr-1with StDev values ranging from 0.0012 mm-1 × sr-1 to 0.0024 mm-1 × sr-1 respectively.The same ROIs used for the quantification of the IBSC were further evaluated with regard to sound attenuation (Figure 6B). ROIs were analyzed in which UCAs were either bound to the matrix or not bound. Each of the fifteen ROIs per condition consisted of nine pixels corresponding to 135 raw radio-frequency (RF) time-signals for the UCA ROIs and similarly fifteen ROIs for the matrix ROIs corresponding to another 135 raw RF time-signals. The frequency-dependent attenuation was calculated over the frequency range from 97 MHz to 104 MHz. Equivalent to our IBSC findings this frequency range consisted of 36 points at a sampling frequency of 400 MHz. The attenuation graphs for the ROIs with bound targeted UCAs and the ROIs with plain matrix over the selected frequency range are shown in Figure 6B. Taken together, detection of targeted bound UCAs is significant compared with unbound UCAs. Our data highlight the potential of targeted UCAs as a means of molecular imaging to detect the early stages of biofilm matrix formation.
In this study, we report for the first time a combined optical and acoustic imaging method of infectious biofilm matrices. Ligand-targeted UCAs were used as a novel method for pre-clinical non-invasive molecular imaging of early to late stage biofilms. These agents were used to target S. aureus biofilm formation and assess the binding efficacy on early to late stage biofilm matrices with respect to their surface area. A combination of acoustic and optical microscopy was used to quantify S. aureus biofilm mechanoelastic properties. We show that time-resolved high-frequency SAM is a viable method for ultrasonic imaging in addition to quantifying mechanical and elastic properties in soft materials (eg. tissues, cells, biofilm matrices) in a non-invasive setting. Moreover, the use of targeted UCAs with high-frequency SAM allow for UCAs detection at higher frequencies other than their resonance frequency. The mechanoelastic properties of the S. aureus biofilm matrix are summarized in Table 1.
Biofilms occurring from infections pose a challenge to current medicine because of the difficulty for early detection and diagnosis. Biofilms protect bacteria and promote resistance to antibiotics and chemotherapeutic agents. Moreover, detecting early and late biofilm formation may be problematic due to their dynamic profile. Individual bacterial cells may detach from the biofilm to colonize other niches or an entire biofilm colony may move as a whole across a region . Thus, biofilm-mediated rippling effects that occur during detachment and transmigration pose biomedical challenges. One example is ventilator-associated pneumonia in immunologically compromised patients that may occur due to biofilm rippling (e.g. cancer patients) . Our data indicates that the binding efficiency of UCAs correlates with matrix biomass. Thus, we propose that the rolling and rippling effects observed during biofilm maturation may reduce biomass and therefore decrease imaging capabilities at the late stages of biofilm matrix formation. It may be that there is a critical timeframe where UCAs bind well and imaging is enhanced and that as the biofilm grows, detaches and ripples, binding is decreased. Developing better detection methodologies and hence diagnostic clinical imaging methods are needed to assess biofilm formation early. By detection of biofilm infections at their earlier stages, this method will potentially offer more treatment options.
Due to the complex structure of biofilm matrices, we focused on the lectins concanavalin A and WGA [5, 42] because biofilms may switch between these two polysaccharides during growth (Table 3). Targeting carbohydrate epitopes that are present in early biofilm matrices may provide novel biofilm markers that will enhance a more optimal molecular imaging, particularly at early stage formation.
Ultrasound imaging devices are readily available in clinical settings and the application of ultrasound techniques for biofilm-relevant infections is familiar to hospital personnel for diseases such as infective endocarditis [12, 23, 43] and cancer . Targeted UCAs have the potential to recognize and bind to early stage biofilm matrix and thus, facilitate an early diagnosis. Our study demonstrated that while more targeted UCAs bound to larger biofilm matrix mass, a significant number of targeted UCAs also bound to less developed biofilm matrices. Targeted UCAs with ultrasound imaging may provide a means for early detection of biofilm formation within a non-invasive setting. This would include detecting endocarditis and biofilm matrix formation at the site of medical implants such as prosthetic devices, and catheters. Moreover, cancer patients rely on catheterization for chemotherapy treatment and biofilms are prevalent at the catheter interface. In immunologically compromised patients, rippling effects of the late stage biofilms have been reported to promote biofilm transmigration to the lungs causing additional complications in treatment [45, 46]. Thus, the use of targeted UCAs may provide a rapid method to facilitate early diagnosis in a number of diseases.
High-frequency ultrasound may be used to assess biofilm development in vitro. Currently clinical biomicroscopy uses frequencies in the range of 15–50 MHz including intravascular ultrasound spectroscopy (IVUS), cardiovascular and ocular applications [47–54]. In particular, clinical imaging applications, such as detecting metastases in the eye, are in the range of 15–50 MHz, although research has been performed to measure at the higher frequency of 75 MHz . The use of higher frequencies in the range of 100 MHz has been used for the imaging of choroidal metastasis . Clinical ultrasound applications have focused on higher frequencies in the range of 100–200 MHz . Low resonance frequencies are used clinically for drug delivery applications in conjunction with UCAs and targeted drug delivery as these low frequencies induce microbubble rupture . Thus, in terms of imaging the higher frequencies provide enhanced imaging capabilities whereas lower resonant frequencies allow for more efficient targeted drug delivery. We report for the first time a method to quantify backscatter intensity and mechanoelastic properties of biofilms [58–64]. With regard to the integrated backscatter and the acoustic attenuation, considering differences in the frequency domain, similar values have been previously reported for cancer cells and tissues [58, 61, 63–66]. A more in-depth understanding of the three-dimensional biofilm matrix structural and mechanoelastic parameters will enhance biofilm imaging and subsequent treatment. Targeted UCAs potentially provide a novel means of imaging for the diagnosis of biofilm infections in vivo.
Materials and methods
Bacterial strains and cultivation of biofilms
We used a penicillin-resistant mutant of S. aureus. S. aureus and coagulase-negative staphylococci account for the majority of device-related infections .
S. aureus cultures were stored frozen at -80°C in 10% glycerol and 90% tryptic soy broth (TSB, T8907, Sigma-Aldrich, St. Louis, USA) solution dissolved in sterile ultrapure water (Alfa Aesar, Ward Hill, MA, USA).
A vial of frozen bacterial culture was thawed at room temperature (RT) and added to 250 mL of TSB. The inoculum was propagated and incubated overnight on an incubator shaker at 37°C and 160 rotations per minute (RPMs). The bacterial cultures were harvested after standardization to an optical density at 600 nm (OD600) of 0.05 relative to the TSB culture medium (Beckman Coulter, Inc., Fullerton, CA, USA).
Biofilm assays were conducted by adding three milliliters of the standardized bacterial culture solution to the pre-treated 35 mm glass (World Precision Instruments, Inc., Sarasota, FL, USA) and polystyrene petri dishes (Greiner Bio-One, Monroe, NC, USA). Glass and polystyrene petri dishes were treated in a previous step with Collagen IV (BD Biosciences, San Jose, CA, USA) for twenty minutes and rinsed in three washing steps with sterile distilled water. Prior to the addition of the inoculum, a 22 × 22 mm sterile micro cover glass (VWR International, LLC, West Chester, PA, USA) was placed into each of the polystyrene petri dishes. The glass and polystyrene petri dishes were then kept inside an incubator shaker at 37°C and 120 RPMs for up to 96 hours without replacement and addition of fresh culture medium in the interim.
Lectins, antibodies and immunofluorescence
Fluorescently-labeled lectins, concanavalin A (conA; binds to α-Man, α-Glc) [5, 42] and wheat germ agglutinin (WGA; binds to (β-GlcNAc)2 and NeuNAc; Sigma-Aldrich Corp., St. Louis, MO, USA) [5, 42] conjugated with FITC were used for the visualization of carbohydrate-containing extracellular polymeric substances in biofilms of S. aureus (Table 3) [5, 42]. Stock solution of ConA at a concentration of 1 mg/mL in 0.1 M sodium bicarbonate (pH 8.3) and WGA at a concentration of 1 mg/mL in phosphate buffered saline (PBS; pH 7.4) were prepared, aliquoted and stored at -20°C. Prior to use, thawed portions of ConA and WGA aliquots were diluted with 0.1 M sodium bicarbonate (pH 8.3) and PBS (pH 7.4) respectively to a lectin final concentration of 10 μg/mL.
The blue-fluorescent nucleic acid stain 4′,6-diamidino-2-phenylindole, dihydrochloride (DAPI, Sigma-Aldrich Corp., St. Louis, MO, USA) was used to visualize bacterial cell distribution in the biofilms. A DAPI stock solution at a concentration of 5 mg/mL and 14.3 mM in ultrapure water was prepared, aliquoted and stored at -20°C. An aliquot was diluted to 300 nM in PBS immediately before use.
The monoclonal immunoglobulin antibody to protein A of S. aureus was used as a conjugation agent for the UCA particles. Anti-Protein A (APA) was developed in rabbit using protein A purified from S. aureus (Sigma-Aldrich Corp., St. Louis, MO, USA). Protein A localizes on the surface of staphylococcal bacterial strains and its distribution is inhomogeneous [68, 69]. The lyophilized content of the vial was reconstituted in 2 mL PBS (pH 7.4) yielding a solution with a protein concentration of 23.7 mg/mL. The lectin from P. aeruginosa (PA-IL, Sigma-Aldrich Corp., St. Louis, MO, USA) was used, similarly to APA, to conjugate the surface of the UCA particles. The lyophilized content was diluted in 1 mL PBS (pH 7.4) yielding a protein concentration of 1 mg/mL. Following the reconstitution of APA and PA-IL, the proteins were biotinylated and conjugated onto the surface of the UCAs according to the method that will be described in more detail.
Sulfo-NHS-LC-Biotin (Thermo Scientific, Rockford, IL, USA) was applied to label APA and PA-IL with biotin. The vial of Sulfo-NHS-LC-Biotin was stored at -20°C and equilibrated to RT before opening to avoid condensation. For the biotin labeling reaction 2.2 mg of Sulfo-NHS-LC-Biotin were dissolved in 400 μL ultrapure water immediately before use yielding a 10 mM solution. A 20-fold molar excess of biotin reagent to label APA and PA-IL, resulting in 4–6 biotin groups per antibody molecule, was found to be suitable. For APA with a concentration of 23.7 mg/mL, a volume of 320 μL Sulfo-NHS-LC-Biotin was used for the biotinylation reaction while 13.5 μL Sulfo-NHS-LC-Biotin were used to label PA-IL with a concentration of 1 mg/mL. Following the incubation on ice for two hours at RT a Zeba® desalt spin column (Thermo Scientific, Rockford, IL, USA) was applied to remove the excess non-reacted and hydrolyzed Sulfo-NHS-LC-Biotin reagent from the APA and PA-IL solutions. The column was placed into a sterile 15 mL falcon tube and centrifuged at 1000 × G for two minutes. After centrifugation the storage buffer collected at the bottom of the falcon tube was discarded, the column placed back into the same falcon tube and equilibrated by adding 2.5 mL of PBS (Thermo Scientific, Rockford, IL, USA) to the top of the resin bed and centrifuging at 1000 × G for two minutes. Next, the flow-through was discarded and the same step was repeated for a total of three times. Subsequently the column was placed into a new sterile 15 mL falcon tube and the antibody solution was applied onto the center of the resin bed. Finally, the column was centrifuged at 1000 × G for two minutes. The collected purified flow-through antibody solutions were aliquoted and stored appropriately.
Targeted ultrasound contrast agents
Biotin-conjugated lipid-encapsulated perfluorocarbon UCAs with a mean diameter of 3.02 μm ± 0.05 μm (Targeson, San Diego, CA, USA) were removed from a sealed vial using a four-way stopcock-syringe combination with a 22 G needle while simultaneously venting the vial with an additional needle. Using a 22 G needle, targestar conjugation buffer (TCB, Targeson, San Diego, CA, USA) was withdrawn into the syringe containing the UCA particles to a total volume of 3.5 mL and centrifuged at 400 × G for three minutes to remove excess free unincorporated lipids from the UCA particle solution. After centrifugation the infranatant was drained drop-wise and the UCAs were re-suspended in 1.0 mL TCB. Afterward, UCAs were incubated with 150 μL FITC-streptavidin (Invitrogen, Carlsbad, CA, USA) at a concentration of 1 mg/mL for twenty minutes at RT with occasional gentle shaking of the vial. The unreacted FITC-streptavidin was removed in a centrifugal washing at 400 × G for three minutes similarly to the previous step and re-suspended in 1.0 mL TCB. Finally, the UCA particles were incubated on ice with either APA or PA-IL for 30 minutes while the unconjugated targeting antibody and ligand molecules were removed by a final centrifugal washing as was described in the previous steps.
Lectin-staining for biofilms
For the S. aureus biofilm matrix, a double staining approach with ConA and WGA was chosen [5, 42]. After incubation for twenty minutes in the dark at RT, excess staining solution was removed by rinsing three times with sterile distilled water.
Epifluorescence microscopy was carried out with a Zeiss Axioskop 2 microscope equipped with an AxioCam MRc Rev 3. Negative and positive controls were conducted using epifluorescence microscopy. For each case four biofilm samples were used to image five random positions on every sample. Each position was imaged by applying the respective filter for FITC, TRITC and DAPI. The negative control did not use any dyes to control for any possible autofluorescence effects.
Ultrasonic investigation of biofilms
Ultrasonic imaging and RF data acquisition was performed with a high-frequency scanning acoustic microscope (Fraunhofer IBMT, St. Ingbert, Germany). A detailed description of the acoustic lens used is shown in Table 2.
The recorded RF data were stored for further processing. The post-processing was conducted using custom-written scripts in MATLAB (The Math-Works, Natick, MA, USA). The scripts were applied for the visual reconstruction in 2D and 3D of the raw RF data for the selected ROIs. RF raw signals were gated by applying a rectangular window function. The window length at 100 MHz excitation center frequency was set such as it corresponded to 10 wavelengths. The wavelength estimation is based on the center frequency of the lens (100 MHz).
The gating of RF raw time-signals allows estimations of scattering properties to be related to distinct ROIs in the volume under interrogation . However, the gating process also allows unwanted frequency content to be added into the backscattered power spectrum which subsequently, leads to inaccurate estimates of scatterer properties. In order to minimize such unwanted effects, a Hamming window was applied . The tapered windows reduced the high-frequency content added into the gated RF time-signals by smoothing the edges.
When the acoustic lens is moved over a ROI where the substrate is covered by an EPS layer under investigation, two echoes are received. One echo originates from the top surface of the layer, S0(t), and the second, S(t), from the interface between the layer and the substrate. These signals can be written as follows [71–74]:
S(t) is the reflected signal from the top and the sample-substrate interface of the EPS matrix. Provided that the defocus is positive meaning that the acoustic lens is elevated above the maximum focus position, it is adequate to constrain the function g to be independent of t and to be a real function of z only. The optimum value of z was found experimentally, by scanning along the z axis and finding the minimum positive value at which the shape of the waveform remained approximately constant as a function of z. The z value was experimentally determined to be 900 μm. Within the approximation of the independence of the waveform shape on z, the signals can be written respectively as :
From the height in amplitude and position on the time axis of each maximum, the following parameters were measured:
where t1 and t2 are the arrival times of the sample and the interface echo respectively (Figure 3A) and t0 is the time arrival of the reference signal (not shown) when no sample is placed in between the acoustic lens and the substrate. The velocity of the coupling medium, which in this case was degassed biofilm medium at 25°C, was approximated to the velocity of distilled water and set to be 1497 m/s [75, 76] while the attenuation of the same medium was set equal to the attenuation of distilled water at 25°C, 2 dB/mm . The density of the medium was calculated with a microbalance and a micropipette at 25°C. From the density of the coupling medium, denoted as ρ cm , and the respective ultrasonic velocity, denoted as v cm , the acoustic impedance, Z cm , of the coupling medium was deduced:
From the difference in time between the reference signal, t0, and the reflection from the sample surface, t1, and by applying the velocity, v0, of the coupling medium, the thickness of the layer is:
From the ratio of magnitude of the reflection A1 from the surface of the layer to the magnitude of the reference signal A0, and by applying the impedance Z0 of the coupling medium as has been calculated in and the acoustic impedance of the substrate Zs, the acoustic impedance of the biofilm sample is:
Finally, from the amplitude A2 of the echo from the interface between the layer and the substrate, the amplitude of the substrate echo A0, the attenuation in the cell, in units of Nepers per unit length, can be calculated as follows:
Integrated backscatter coefficient
Phosphate buffered saline
Region of interest
Revolutions per minute
Targeson conjugation buffer
Tryptic soy broth.
Liu D, Lau YL, Chau YK, Pacepavicius G: Simple technique for estimation of biofilm accumulation. Bull Environ Contam Toxicol. 1994, 53 (6): 913-918.
Allison DG, Ruiz B, SanJose C, Jaspe A, Gilbert P: Extracellular products as mediators of the formation and detachment of Pseudomonas fluorescens biofilms. Fems Microbiol Letters. 1998, 167 (2): 179-184.
Costerton JW, Stewart PS, Greenberg EP: Bacterial biofilms: a common cause of persistent infections. Science. 1999, 284 (5418): 1318-1322.
Wingender J, Neu TR, Flemming HC: What are Bacterial Extracellular Substances?. Microbial Extracellular Polymeric Substances: Characterization, Structure and Function. 1st edition. Edited by: Wingender J, Neu TR, Flemming HC. Berlin: Springer, 1999, 1-19.
Strathmann M, Wingender J, Flemming HC: Application of fluorescently labelled lectins for the visualization and biochemical characterization of polysaccharides in biofilms of pseudomonas aeruginosa. J Microbiol Methods. 2002, 50 (3): 237-248.
Hall-Stoodley L, Costerton JW, Stoodley P: Bacterial biofilms: from the natural environment to infectious diseases. Nat Rev Microbiol. 2004, 2 (2): 95-108.
Davey ME, GA O’T: Microbial biofilms: from ecology to molecular genetics. Microbiol Mol Biol Rev. 2000, 64 (4): 847-867.
Flemming HC, Wingender J: The biofilm matrix. Nat Rev Microbiol. 2010, 8 (9): 623-633.
Lambe DW, Ferguson KP, Mayberry-Carson KJ, Tober-Meyer B, Costerton JW: Foreign-body-associated experimental osteomyelitis induced with bacteroides fragilis and staphylococcus epidermidis in rabbits. Clin Orthop Relat Res. 1991, 266: 285-294.
Harris LG, Richards RG: Staphylococci and implant surfaces: a review. Injury-Int J Care Injured. 2006, 37: 3-14.
Baldassarri L, Montanaro L, Creti R, Arciola CR: Underestimated collateral effects of antibiotic therapy in prosthesis-associated bacterial infections. Int J Artif Organs. 2007, 30: 786-791.
Kaplan JB: Methods for the treatment and opinion prevention of bacterial biofilms. Expert Opinion Therap Patents. 2005, 15 (8): 955-965.
Sullam PM, Drake TA, Sande MA: Pathogenesis of endocarditis. Am J Med. 1985, 78 (6B): 110-115.
Caldwell DA, Lovasik D: Endocarditis in the immunocompromised. Am J Nurs. 2002, 32-36. Suppl
Donlan RM, Costerton JW: Biofilms: survival mechanisms of clinically relevant microorganisms. Clin Microbiol Rev. 2002, 15 (2): 167-193.
Petti CA, Fowler VG: Staphylococcus aureus bacteremia and endocarditis. Cardiol Clin. 2003, 21 (2): 219-233. vii
Nickel JC, Wright JB, Ruseska I, Marrie TJ, Whitfield C, Costerton JW: Antibiotic-resistance of pseudomonas-aeruginosa colonizing a urinary catheter invitro. Eur J Clin Microbiol Infect Dis. 1985, 4 (2): 213-218.
Allison DG, Gilbert P: Modification by surface association of antimicrobial susceptibility of bacterial-populations. J Ind Microbiol. 1995, 15 (4): 311-317.
Stewart PS, Costerton JW: Antibiotic resistance of bacteria in biofilms. Lancet. 2001, 358 (9276): 135-138.
Parsek MR, Singh PK: Bacterial biofilms: an emerging link to disease pathogenesis. Annu Rev Microbiol. 2003, 57: 677-701.
Furuya EY, Lowy FD: Antimicrobial strategies for the prevention and treatment of cardiovascular infections. Curr Opin Pharmacol. 2003, 3 (5): 464-469.
Durack DT, Lukes AS, Bright DK, Alberts MJ, Bashore TM, Corey GR, Douglas JM, Gray L, Harrell FE, Harrison JK, Heinle SA, Morris A, Kisslo JA, Nicely LM, Oldham N, Penning LM, Sexton DJ, Towns M, Waugh RA: New criteria for diagnosis of infective endocarditis - utilization of specific echocardiographic findings. Am J Med. 1994, 96 (3): 200-209.
Baddour LM, Bettmann MA, Bolger AF, Epstein AE, Ferrieri P, Gerber MA, Gewitz MH, Jacobs AK, Levison ME, Newburger JW, Pallasch TJ, Wilson WR, Baltimore RS, Falace DA, Shulman ST, Tani LY, Taubert KA: Infective endocarditis: diagnosis, antimicrobial therapy, and management of complications. Circulation. 2005, 112 (15): 2373-
Shemesh H, Goertz DE, van der Sluis LW, de Jong N, Wu MK, Wesselink PR: High frequency ultrasound imaging of a single-species biofilm. J Dent. 2007, 35 (8): 673-678.
Vaidya K, Osgood R, Ren D, Pichichero ME, Helguera M: Ultrasound imaging and characterization of biofilms based on wavelet de-noised radiofrequency data. Ultrasound Med Biol. 2014, 40 (3): 583-595.
Good MS, Wend CF, Bond LJ, McLean JS, Panetta PD, Ahmed S, Crawford SL, Daly DS: An estimate of biofilm properties using an acoustic microscope. IEEE Trans Ultrason Ferroelectr Freq Control. 2006, 53 (9): 1637-1646.
Holmes AK, Laybourn-Parry J, Parry JD, Unwin ME, Challis RE: Ultrasonic imaging of biofilms utilizing echoes from the biofilm/air interface. IEEE Trans Ultrason Ferroelectr Freq Control. 2006, 53 (1): 185-192.
Kujundzic E, Fonseca AC, Evans EA, Peterson M, Greenberg AR, Hernandez M: Ultrasonic monitoring of early-stage biofilm growth on polymeric surfaces. J Microbiol Methods. 2007, 68 (3): 458-467.
Sbeity F, Menigot S, Charara J, Girault JM: Contrast improvement in sub- and ultraharmonic ultrasound contrast imaging by combining several hammerstein models. Int J Biomed Imag. 2013, 2013: 270523-
Anderson CR, Hu X, Zhang H, Tlaxca J, Decleves A-E, Houghtaling R, Sharma K, Lawrence M, Ferrara KW, Rychak JJ: Ultrasound molecular imaging of tumor angiogenesis with an integrin targeted microbubble contrast agent. Investig Radiol. 2011, 46 (4): 215-224.
Klibanov AL: Microbubble contrast agents - targeted ultrasound imaging and ultrasound-assisted drug-delivery applications. Investig Radiol. 2006, 41 (3): 354-362.
Klibanov AL: Ultrasound molecular imaging with targeted microbubble contrast agents. J Nucl Cardiol. 2007, 14 (6): 876-884.
Klibanov AL: Preparation of targeted microbubbles: ultrasound contrast agents for molecular imaging. Med Biol Eng Comput. 2009, 47 (8): 875-882.
Klibanov AL, Hughes MS, Villanueva FS, Jankowski RJ, Wagner WR, Wojdyla JK, Wible JH, Brandenburger GH: Targeting and ultrasound imaging of microbubble-based contrast agents. Magnet Reson Mat Biol Physics Med. 1999, 8 (3): 177-184.
Unnikrishnan S, Klibanov AL: Microbubbles as ultrasound contrast agents for molecular imaging: preparation and application. Am J Roentgenol. 2012, 199 (2): 292-299.
Weiss EC, Anastasiadis P, Pilarczyk G, Lernor RM, Zinin PV: Mechanical properties of single cells by high-frequency time-resolved acoustic microscopy. Ultrason Ferroelectr Frequen Contr IEEE Trans. 2007, 54 (11): 2257-2271.
Weiss EC, Lemor RM, Pilarczyk G, Anastasiadis P, Zinin PV: Imaging of focal contacts of chicken heart muscle cells by high-frequency acoustic microscopy. Ultrasound Med Biol. 2007, 33 (8): 1320-1326.
Zinin PV, Allen JS: Deformation of biological cells in the acoustic field of an oscillating bubble. Phys Rev E. 2009, 79 (2 Pt 1): 021910-
Zinin PV, Allen JS, Levin VM: Mechanical resonances of bacteria cells. Phys Rev E Stat Nonlin Soft Matter Phys. 2005, 72 (6 Pt 1): 061907-
Moran CM, Watson RJ, Fox KAA, McDicken WN: In vitro acoustic characterisation of four intravenous ultrasonic contrast agents at 30 MHz. Ultrasound Med Biol. 2002, 28 (6): 785-791.
Inglis TJ: Evidence for dynamic phenomena in residual tracheal tube biofilm. Br J Anaesth. 1993, 70 (1): 22-24.
Goldstein IJ, Hayes CE: The lectins: carbohydrate-binding proteins of plants and animals. Adv Carbohydr Chem Biochem. 1978, 35: 127-340.
Archibald LK, Gaynes RP: Hospital-acquired infections in the United States. The importance of interhospital comparisons. Infect Dis Clin North Am. 1997, 11 (2): 245-255.
Azzopardi EA, Ferguson EL, Thomas DW: The enhanced permeability retention effect: a new paradigm for drug targeting in infection. J Antimicrob Chemother. 2013, 68 (2): 257-274.
Inglis TJJ, Titmeng L, Mahlee N, Eekoon T, Kokpheng H: Structural features of tracheal tube biofilm formed during prolonged mechanical ventilation. Chest. 1995, 108 (4): 1049-1052.
Al Akhrass F, Al Wohoush I, Chaftari AM, Reitzel R, Jiang Y, Ghannoum M, Tarrand J, Hachem R, Raad I: Rhodococcus bacteremia in cancer patients is mostly catheter related and associated with biofilm formation. Plos ONE. 2012, 7 (3): e32945-
Hejsek L, Pasta J: Ultrasonographic biomicroscopy of the eye. Ultrazvukova? Biomikroskopie oka. 2005, 61 (4): 273-280.
Liang HD, Blomley MJK: The role of ultrasound in molecular imaging. Br J Radiol. 2003, 76 (suppl_2): S140-S150.
Silverman RH: High-resolution ultrasound imaging of the eye - a review. Clinical Exper Ophthalmol. 2009, 37 (1): 54-67.
Silverman RH, Cannata J, Shung KK, Gal O, Patel M, Lloyd HO, Feleppa EJ, Coleman DJ: 75 MHz ultrasound biomicroscopy of anterior segment of eye. Ultrason Imaging. 2006, 28 (3): 179-188.
Snetkova YV, Levin VM, Petriniuk YS, Denisov AF, Bogachenkov AN, Denisova LA, Khramtzova YA: Application of the method of acoustic microscopy for the study of eye tissues. Morfologiya. 2005, 127 (2): 72-75.
Goertz DE, Frijlink ME, De Jong N, Steen A: Nonlinear intravascular ultrasound contrast imaging. Ultrasound Med Biol. 2006, 32 (4): 491-502.
Goertz DE, Frijlink ME, Tempel D, Bhagwandas V, Gisolf A, Krams R, de Jong N, van der Steen AFW: Subharmonic contrast intravascular ultrasound for vasa vasorum imaging. Ultrasound Med Biol. 2007, 33 (12): 1859-1872.
Phillips LC, Klibanov AL, Wamhoff BR, Hossack JA: Intravascular ultrasound detection and delivery of molecularly targeted microbubbles for gene delivery. IEEE Trans Ultrason Ferroelectr Freq Control. 2012, 59 (7): 1596-1601.
Witkin AJ, Fischer DH, Shields CL, Reichstein D, Shields JA: Enhanced depth imaging spectral-domain optical coherence tomography of a subtle choroidal metastasis. Eye. 2012, 26 (12): 1598-1599.
Knspik DA, Starkoski B, Pavlin CJ, Foster FS: A 100–200 MHz ultrasound biomicroscope. IEEE Trans Ultrason Ferroelectr Freq Control. 2000, 47 (6): 1540-1549.
Liu Y, Miyoshi H, Nakamura M: Encapsulated ultrasound microbubbles: therapeutic application in drug/gene delivery. J Control Release. 2006, 114 (1): 89-99.
Saijo Y, Sasaki H, Okawai H, Nitta S-I, Tanaka M: Acoustic properties of atherosclerosis of human aorta obtained with high-frequency ultrasound. Ultrasound Med Biol. 1998, 24 (7): 1061-1064.
Saijo Y, Jorgensen CS, Falk E: Ultrasonic tissue characterization of collagen in lipid-rich plaques in apoE-deficient mice. Atherosclerosis. 2001, 158 (2): 289-295.
Saijo Y, Santos E, Sasaki H, Yambe T, Tanaka M, Hozumi N, Kobayashi K, Okada N: Ultrasonic tissue characterization of atherosclerosis by a speed-of-sound microscanning system. IEEE Trans Ultrason Ferroelectr Frequen Contr. 2007, 54 (8): 1571-1577.
Saijo Y, Miyakawa T, Sasaki H, Tanaka M, Nitta SI: Acoustic properties of aortic aneurysm obtained with scanning acoustic microscopy. Ultrasonics. 2004, 42 (1–9): 695-698.
Brand S, Solanki B, Foster DB, Czarnota GJ, Kolios MC: Monitoring of cell death in epithelial cells using high frequency ultrasound spectroscopy. Ultrasound Med Biol. 2009, 35 (3): 482-493.
Brand S, Weiss EC, Lemor RM, Kolios MC: High frequency ultrasound tissue characterization and acoustic microscopy of intracellular changes. Ultrasound Med Biol. 2008, 34 (9): 1396-1407.
Strohm EM, Czarnota GJ, Kolios MC: Quantitative measurements of apoptotic cell properties using acoustic microscopy. IEEE Transactions Ultrason Ferroelectr Frequen Contr. 2010, 57 (10): 2293-2304.
Baddour RE, Sherar MD, Hunt JW, Czarnota GJ, Kolios MC: High-frequency ultrasound scattering from microspheres and single cells. J Acoust Soc Am. 2005, 117 (2): 934-943.
Saijo Y, Tanaka M, Okawai H, Sasaki H, Nitta SI, Dunn F: Ultrasonic tissue characterization of infarcted myocardium by scanning acoustic microscopy. Ultrasound Med Biol. 1997, 23 (1): 77-85.
Baddour LM, Bettmann MA, Bolger AF, Epstein AE, Ferrieri P, Gerber MA, Gewitz MH, Jacobs AK, Levison ME, Newburger JW, Pallasch TJ, Wilson WR, Baltimore RS, Falace DA, Shulman ST, Tani LY, Taubert KA: Nonvalvular cardiovascular device-related infections. Circulation. 2003, 108 (16): 2015-2031.
DeDent AC, McAdow M, Schneewind O: Distribution of protein A on the surface of staphylococcus aureus. J Bacteriol. 2007, 189 (12): 4473-4484.
Schneewind O, Fowler A, Faull KF: Structure of the cell wall anchor of surface proteins in staphylococcus aureus. Science. 1995, 268 (5207): 103-106.
Oelze ML, O’Brien WD: Improved scatterer property estimates from ultrasound backscatter for small gate lengths using a gate-edge correction factor. J Acoust Soc Am. 2004, 116 (5): 3212-3223.
Briggs A: Advances in Acoustic Microscopy. New York: Plenum Press, 1995.
Briggs A, Kolosov O: Acoustic Microscopy. New York: Clarendon, 2010.
Briggs GAD, Rowe JM, Sinton AM, Spencer DS: Quantitative Methods in Acoustic Microscopy. Ultrasonics Symposium Proceedings: 1988. Chicago, IL, USA: Publ by IEEE, 1988, 743-749.
Briggs GAD, Wang J, Gundle R: Quantitative acoustic microscopy of individual living human cells. J Microsc. 1993, 172 (Pt1): 3-12.
Martin G, Carroll ET: Tables of the speed of sound in water. J Acoustic Soc Am. 1959, 31 (1): 75-76.
Greenspan M, Tschiegg CE: Tables of the speed of sound in water. J Acoustic Soc Am. 1959, 31 (1): 75-76.
Akashi N, Kushibiki J, Dunn F: Acoustic properties of egg yolk and albumen range 20–400 MHz. J Acoust Soc Am. 1997, 102 (6): 3774-3778.
We thank Dr. Tung T. Hoang (Microbiology Dept., University of Hawaii at Manoa) for providing us with the bacterial strain. We would like to thank Dr. Terry Matsunaga (Dept. of Radiology, University of Arizona, Tucson, AZ, USA), Dr. Joshua Rychak (Targeson, San Diego, CA, USA) and Dr. Alexander L. Klibanov (Dept. of Biomedical Engineering, University of Virginia, Charlottesville, VA, USA) for sharing their technical expertise with us and for valuable discussions. This work was supported by: National Institutes of Health (NCRR P20-RR016453 and RO1GM104984 to M.L.M).
The authors declare that they have no competing interests.
PA, KDAM, JSA, MLM: Designed the study. PA, KDAM: Conducted the experiments, performed data analysis, performed statistical analysis. PA, MLM: Prepared and edited the manuscript. JSA, KDAM: Edited the manuscript. All authors added intellectual content, read and approved the final version.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made.
The images or other third party material in this article are included in the article’s Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder.
To view a copy of this licence, visit https://creativecommons.org/licenses/by/4.0/.
The Creative Commons Public Domain Dedication waiver (https://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated in a credit line to the data.
About this article
Cite this article
Anastasiadis, P., Mojica, K.D.A., Allen, J.S. et al. Detection and quantification of bacterial biofilms combining high-frequency acoustic microscopy and targeted lipid microparticles. J Nanobiotechnol 12, 24 (2014). https://doi.org/10.1186/1477-3155-12-24
- Targeted therapy
- Lipid particles
- Biofilm matrix
- Targeted ultrasound contrast agents
- Acoustic microscopy
- Molecular imaging