- Open Access
Recent progress in advanced biomaterials for long-acting reversible contraception
Journal of Nanobiotechnology volume 20, Article number: 138 (2022)
Unintended pregnancy is a global issue with serious ramifications for women, their families, and society, including abortion, infertility, and maternal death. Although existing contraceptive strategies have been widely used in people's lives, there have not been satisfactory feedbacks due to low contraceptive efficacy and related side effects (e.g., decreased sexuality, menstrual cycle disorder, and even lifelong infertility). In recent years, biomaterials-based long-acting reversible contraception has received increasing attention from the viewpoint of fundamental research and practical applications mainly owing to improved delivery routes and controlled drug delivery. This review summarizes recent progress in advanced biomaterials for long-acting reversible contraception via various delivery routes, including subcutaneous implant, transdermal patch, oral administration, vaginal ring, intrauterine device, fallopian tube occlusion, vas deferens contraception, and Intravenous administration. In addition, biomaterials, especially nanomaterials, still need to be improved and prospects for the future in contraception are mentioned.
The importance of sexual life to human health and the overall well-being of society has been increasingly recognized. Proper sex life not only relieves individual stress and maintains family harmony, but also maintains the female reproductive organ health . However, unintended pregnancy is a global issue with serious ramifications for women, their families, and society, including abortion, infertility, and maternal death. There are approximately 85 million unwanted pregnancies and 40–60 million abortions worldwide each year . Abortion inevitably damages the female endometrium, which can cause menstrual disorder, uterine adhesion, intrauterine infection, habitual abortion, and even lifelong infertility . Although existing contraceptive strategies have been widely used in people’s lives, there has not been much satisfactory feedback. Long-term oral hormonal contraceptives can cause endocrine disorder, impaired glucose tolerance, abnormal liver function, and even an increased risk of thrombotic disease [4, 5]. The improper use of condoms can lead to contraceptive failure, and allergic symptoms may be caused by the application of latex . According to the report from the World Health Organization (WHO) to investigate the effectiveness of contraceptive methods in 2020, traditional therapies e.g., sperm–egg fusion disruption alone, emergency drug contraception, and standard date calculation are all not effective in contraception (Table 1) . In addition, although both male and female sterilization have low rates of re-pregnancy, they do not meet the requirements of post-contraceptive reproduction and protection against the risk of future child demise. Therefore, the design and development of long-acting reversible contraception are emerging as a promising strategy to address the issues above.
In recent years, biomaterials-based long-acting reversible contraception has received increasing attention from the viewpoint of fundamental research and practical applications mainly owing to improved delivery routes and controlled drug delivery [8,9,10]. “Biomaterials are those materials—be it natural or synthetic, alive or lifeless, and usually made of multiple components—that has been engineered to interact with biological systems for a medical purpose—either a therapeutic (treat, augment, repair, or replace a tissue function of the body) or a diagnostic one.” [11, 12]. As shown in Table 2, various types of biomaterials have been used in a wide range of contraceptives based on their characteristics and superiorities. And the report of scientific researches showed that biomaterials are widely studied in the field of contraception in the form of delivery carriers compared to oral or injection administration (Fig. 1). Biomaterials with various components and architectures provide visible and tangible reactions to modulate tissue microenvironment for practical applications [13,14,15,16,17,18,19,20,21,22]. Particularly, for long-acting reversible contraceptives, biomaterials can not only serve as an effective drug or hormonal carrier for long-term stable release in vivo but also respond to conditioned stimuli in the body, solving problems such as peristalsis causing the device to slip. In addition, the unique (bio)physicochemical properties of biomaterials including spermicidal action, stable degradability, and stiffness, can also be used directly or as an adjunct to long-acting contraception. Also, the controllable biodegradability, photothermal and chemical responses of the advanced biomaterials can be used to achieve a reversible recovery of fertility with minimal or no damage.
In this review, we summarized recent advances in biomaterials-based long-acting reversible contraception via various delivery routes, including subcutaneous implant, transdermal patch, oral administration, vaginal ring, intrauterine device, fallopian tube occlusion, vas deferens contraception, and intravenous administration (Fig. 2). In addition, biomaterials, especially nanomaterials, still need to be improved and prospects for the future in contraception are mentioned.
Reproductive system and fertilization process
The reproductive system refers to the tissues, glands, and organs required for the reproduction of human offspring. Regulated by the neuro-endocrine system, the system is responsible for the maturation of sperm and eggs, fertilization, pregnancy, and childbirth . The main contraceptive methods intervene in the early stages of the pregnancy by inhibiting the secretion of hormones or cytokines, blocking pathways, and altering the environment of implantation in the uterus, which prevents the maturation of the sperm and egg, cuts off the sperm–egg fusion pathway and interfere with the implantation of the fertilized egg .
Male reproductive system and spermatogenesis
A male reproductive system is a group of organs that can promote sperm maturation, storage, and ejaculation, as well as secrete androgens for development and maintain androgynous characteristics . It consists mainly of the gonads (testis), the vas deferens (epididymis, vas deferens, ejaculatory ducts, and urethra), the accessory glands (seminal vesicle glands, prostate gland, urethral bulb glands), the scrotum, and the penis.
The testis is the main gland of spermatogenesis and androgen secretion, which are regulated by the hypothalamic–pituitary–gonadal axis (Fig. 3A) . The hypothalamus secretes gonadotropin-releasing hormone (GnRH) into the hypothalamic–pituitary portal system, stimulating the anterior pituitary gland, which further promotes the release of luteinizing hormone (LH) and follicle-stimulating hormone (FSH). LH stimulates the secretion of carbon chain enzymes from testicular interstitial cells to convert cholesterol into testosterone, and FSH stimulates testicular supporting cells to promote spermatogenesis as well as release inhibin B and Mullerian inhibiting substances (MIS) hormone. Testosterone is essential for maintaining a normal libido and sexual function, promoting the maturation of the sexual organs, and maintaining the secondary sexual characteristics of the male.
The primitive spermatocytes are located in the periphery of the spermatogenic tubules and the supporting cells activated by FSH promote spermatogenesis. During 13 days, the spermatogonia complete two meiotic divisions to form haploid spermatocytes, which are simultaneously released into the epididymis for maturation and storage in the caudal epididymis until ejaculation occurs. On the other hand, testosterone reduces the secretion of LH and FSH through negative feedback from the hypothalamus and anterior pituitary. Inhibin B reduces the release of FSH by a similar negative feedback mechanism.
During sexual arousal, the penis is engorged with blood causing penis erection. With the sexual stimulation continuing, the smooth muscle of the epididymis contracts to push sperm into the vas deferens, which are located in the spermatic cord. The vas deferens are connected to the seminiferous ducts near the prostate and carry the sperm to the ejaculatory ducts to reach the female vagina.
Female reproductive system and fertilization
The female reproductive system plays a critical role in ovulation, fertilization, pregnancy, childbirth, promoting growth, as well as developing and maintaining female secondary sexual characteristics . It consists mainly of the internal genitalia, including the vagina, uterus, fallopian tubes and ovaries, and the vulva. Unlike the male reproductive system, the female reproductive system has a regular menstrual cycle and ovulation regulated by the hypothalamic–pituitary–gonadal axis (Fig. 3B).
The menstrual cycle refers to the cyclical changes in the functional layer of the endometrium under the cyclical action of estrogen and androgens secreted by the ovary from puberty, and the process of endometrial exfoliation, bleeding, repair and proliferation occurs every 28 days, including the follicular phase, ovulation phase, and luteal phase. The ovulation period generally occurs approximately 14 days before the next menstrual cramp. Similarly, GnRH secreted by the hypothalamus stimulates the anterior pituitary to release FSH and LH. Under the stimulation of FSH, the granulosa cells in the ovary secrete estradiol (E2) and inhibin, and the negative feedback regulates the secretion of the hypothalamus and pituitary gland so that the level of FSH gradually decreases. LH acts on the surface receptors of the follicular membrane cells to promote follicular development and secrete androgens. In the late follicular phase, with the E2 level continuing to rise, it produces a positive feedback effect on the hypothalamus and pituitary, which causes the LH level to rise sharply, and ovulation occurs. Following the releasing its ovum, the follicle transforms into a corpus luteum and secretes progesterone, 17β-progesterone, oestradiol, androstenedione, etc. Progesterone stimulates the transformation of the endometrium from proliferative to secretory, which involves the secretion of chemokines, growth factors, and cell adhesion molecules, the thickening of the endometrium, and the distribution of spiral arteries that facilitate the implantation of the zygote.
The sperm and egg fusion in the abdomen of the fallopian tube and complete the fertilization process to form a zygote. The zygote remains in the fallopian tube for 3 days and undergoes a series of cell divisions, differentiation, and eventually the formation of a mulberry embryo. The mulberry embryo moves along the fallopian tube towards the horn of the uterus and enters the uterus in the form of an embryo. Over approximately 6 days, the zygote develops into a blastocyst and completes its implantation in the endometrium, while actively secreting human chorionic gonadotropin (hCG), which allows the corpus luteum to continue to produce progesterone, thus maintaining the thickened endometrium for the continuation of the pregnancy.
Advanced biomaterials for contraception by various routes
The choice of contraceptive method has always been a focus of concern, with safety, effectiveness, availability (including accessibility and affordability) and acceptability being factors often considered . It is noted that no form is 100% effective, but certain methods can be particularly effective through contraceptive mechanisms. In addition to preventing pregnancy, some contraceptive methods also have effects like preventing sexually transmitted infections and antibacterial properties . The frequently used methods have been listed along with their mechanisms and pros and cons analysis, as shown in Table 3. It has been demonstrated that biomaterials with unique (bio)physicochemical characteristics can not only improve the effectiveness and safety of contraceptive methods through a series of routes, but also focus on exploring post-contraceptive reversibility and allowing long-acting reversible contraception.
Subcutaneous implantation of biomaterials-based contraceptives is a combination of progestogen and biomaterials with different structures (e.g., capsules or small rods) that are implanted under the skin and released the drug in vivo steady and constantly. They have a long-acting period and great reversibility. Single implantation can last up to 5 years with a pearl index of 0.1 and full fertility recovery is expected after removal of the implant .
In the 1960s, the Population Council developed the first subcutaneous implanted contraceptive, levonorgestrel silicone rods (Norplant®), which consisted of six capsules containing 216 mg of levonorgestrel (LNG) and releasing 40–50 μg of LNG per day . LNG, as a type of progesterone, works on the hypothalamus and pituitary gland, inhibiting the secretion of FSH and LH, which prevents ovulation from being completed, and has significant anti-estrogenic activity . In addition, LNG thickens the cervical mucus and hinders sperm penetration, and can cause the endometrium to thin, which is not conducive to embryo implantation . Although silicone rubber possesses excellent stability, the implant is still considered to be a foreign object in vivo and must be removed when it expires . Moreover, there may be individual differences in the reaction of the human body to silicone rubber, resulting in significant aging of the silicone tubing and difficulty in removal. Jadelle® is a second-generation product, which contains 150 mg of LNG in two silicone rods and is effective for 5 years . Its contraceptive effect is similar to that of Norplant®, but the difficulty of removal is greatly reduced by the reduction in the number of implants. Implanon® with ethyl acetate (EVA) used as the carrier matrix is another type of implantable contraceptive extended-release delivery system. EVA not only has good elasticity, flexibility, water resistance, corrosion resistance, but is more tolerant to fillers than silicone rubber. The subcutaneous administration was last for 3 years and the Pearl index was 0.08 . In addition, Nexplanon® is a modified version of Implanon®, and the addition of BaSO4 allows the use of X-ray-assisted positioning during the operation .
The biodegradable implant has been studied to avoid secondary trauma to the patients when the device needs to be removed surgically after its expiration. It is delivered through particles loaded with steroids, which are able to ensure both long-term stability with the drug and constant release in biodegradable form. However, both the polymer matrix and the biodegradation products must be innocuous and not accumulate in the body. As a hydrophobic degradable biomaterial, polylactic acid (PLA) is not only capable of controlled degradation but also possesses excellent mechanical and physical properties, tensile resistance, and low biological toxicity PLA forms particles. The releasing system formulated by POE with Na2CO3 and hydrophobic steroids can prevent liquid in vivo from entering the polymer inside under the prolonged body fluid erosion . Additionally, POE can be hydrolyzed in an acidic environment to release steroids in a time-predictive manner . Meanwhile, it has good biocompatibility as a matrix material, which can be used without causing tissue necrosis or systemic immune reaction. The subcutaneous implant prepared by POE and norethisterone (NET) was pharmacologically studied in baboons and showed that serum levels reached a steady state within 2 days after insertion and remained relatively stable for 5 months . However, local irritation, mild to moderate connective tissue granulomas, and inflammatory reactions were observed in preclinical studies.
As a hydrophobic degradable biomaterial, polylactic acid (PLA) is not only capable of controlled degradation but also possesses excellent mechanical and physical properties, tensile resistance, and low biological toxicity . PLA forms particles with the drug, which are stable in the body, and release the drug at a constant rate. However, degradation of PLA in vivo generates CO2 and H2O, which can locally produce an acidic environment with a certain probability of causing inflammation . Copolymers prepared from 90% PLA and 10% polyglycolic acid (PGA) incorporate 20% NET to form 90 to 180 μm particles and achieve zero-level kinetic release . Clinical trials demonstrated that 90 days after the operation, the average serum NET levels ranged from 1 to 3 ng/mL. Ovulation was inhibited in all subjects, and NET serum concentrations decreased significantly after 100 days .
Compared with biodegradable biomaterials such as POE and PLA which are still in the research and development stage, poly-(ε-caprolactone) (PCL) has been developed for its unique physicochemical properties and its related subcutaneous contraceptive implant is available commercially, called Capronor® . During subcutaneous implantation, the ester bonds of PCL sustain cleavage but maintain the physical integrity until the drug supply is depleted . Further, PCL is degraded to 3000 molecular weight oligomers and finally degraded to carbon dioxide and water . The high permeability of PCL and its copolymers has also been considered as an excellent indicator of drug release diffusion of subcutaneous microcapsules. Feedbacks in clinical phase I evaluation of Capronor® varied significantly between participants, but plasma LNG levels typically remained constant throughout an observation cycle, with average plasma concentrations ranging from 450 to 650 pg/mL . However, cases of interrupted menstrual bleeding patterns occurred in the first 3 months after insertion .
The transdermal patch refers to a flake-like preparation made of drugs and biomaterials that can be adhered to the skin and produce systemic or local effects . The drug can be delivered to the skin through the surface layer of human skin non-invasively . Compared to oral administration or subcutaneous implantation, transdermal patches can avoid incompatibility of the drug with the gastrointestinal tract or first-pass metabolism. In addition, the sustained drug release in the patch maintains a stable effective blood concentration with reduced side effects associated with burst release. Besides, the frequency of patchoperation is greatly reduced compared to oral and injectable administration, as well as being self-administered by the patient, which greatly improves adaptability and acceptability. Furthermore, the characteristics of the material itself are also crucial for patch application, e.g., homogeneity and release with the drug, adhesion and permeability to the skin, microbiological limits and the degree of drug residue.
Ortho Evra® is the first transdermal contraceptive patch to be marketed, and it was approved by the U.S. Food and Drug Administration (FDA) in 2001 . The patch consists of three layers. The backing layer is composed of a beige flexible film made up of a low-density pigmented polyethylene outer layer and a polyester inner layer. It provides structural support and protection. The middle layer contains polyisobutylene/polybutene adhesive, crospovidone, non-woven polyester fabric, and lauryl lactate as inactive components. The active components in this layer include 0.75 mg of ethinyl estradiol and 6 mg of methyl progesterone, which release ~ 35 μg of ethinylestradiol and 150 μg of norgestimate per day. The third layer is the release liner, which protects the adhesive layer during storage and is removed just before the application . The patch is used continuously for 1 week, and its compliance is greatly improved compared to the oral contraceptive once a day. The effect of avoiding pregnancy is satisfactory, and the Perl index in clinical experiments is only 1.07 .
Dissolving microneedle (DMN)
Recently, microneedle has been developed as a type of micron-sized drug delivery channel into the skin, allowing large molecules to pass through and only puncture the stratum corneum without touching the nerve endings, so there is slight pain or even no pain [61, 62]. DMN is composed of biocompatible and biodegradable polymers that degrade in the body, dissolving in the skin at a constant rate and releasing the encapsulated drug gradually, leaving no acrid or harmful biological waste . However, there are still some hardships to be solved that is how to promote the dissolving ability and drug loading proportion. The poor dissolving ability of the carrier always leads to inefficient drug release and potential skin irritation. Thermosensitive materials containing CS and β-GP can improve the dissolving ability of DMN by temperature-induced phase transition . In addition, hydroxypropyl beta-cyclodextrin (HP-β-CD) enhances the solubility of low water-soluble steroids by forming an encapsulated compound, which increases significantly the limited drug loading . Yao et al. presented a novel type of DMNs with great mechanical strength, dissolving ability, and drug release profile in vitro . LNG was encapsulated into HP-β-CD to form inclusion compounds, while the DMN was prepared with the inclusion compounds and thermosensitive materials containing chitosan and β-GP. The DMN has successfully penetrated the skin without skin irritation, and the DMN was dissolved 69.32 ± 4.23% within 2 h, which was almost twice the dissolution rate of conventional DMN. In addition, HP-β-CD loaded with LNG increased the drug loading capacity to 3380.56 μg/mL, exhibiting a similar pharmacokinetic curvature as oral LNG suspensions (Fig. 4). PLA and polylactic acid co-glycolic acid (PLGA) were chosen as microneedle materials since these biodegradable polymers offered the advantages of biocompatibility and mechanical properties, with LNG loading over 40% and sustained slow release for weeks to 11 months [64, 65]. Li et al. reported a microneedle patch design with degradable PLA and PLGA on the application of sustained-release LNG. (Fig. 5) . The separated microneedles were then slowly biodegraded in the skin to control the constant release and systematic delivery of encapsulated LNG for more than a month. In the rats model, microneedle patches were well tolerated with few signs of use and maintained plasma hormone concentrations above human therapeutic levels within a month.
To address the rapid separation between the microneedle and patch backing, Li and co-workers developed an effervescent backing composed of sodium bicarbonate and citric acid, which can react to the interstitial skin fluid from solid to the gas state, achieving effective separation of the microneedle in less than 1 min (Fig. 5) . Simple and rapid separation can greatly reduce the patient’s degree of sense of pain during detachment, the risk of skin damage, and the loss of active microneedle components. Similar phenomena were also observed in the bubble structure and porous structure between the microneedle and the backing [66, 67]. Li et al. reported that the bubble substrate formed by water-soluble polyvinyl alcohol and sucrose has strong compression resistance but weak shear resistance without changing the size and shape of the microneedle, which can achieve rapid separation after being inserted into the skin . In addition, Lee et al. proposed that the porous structure of the platform pad was designed to provide sufficient pressure support for complete insertion into the skin and rapid separation under shear force during detachment in less than 1 s . Polyvinylpyrrolidone (PVP) is considered to be an ideal material for the preparation of porous structures, which can be optimized for porosity and mechanical properties by adjusting the PVP/H2O ratio . As a matrix carrier of the microneedle, PVP with great biocompatibility and high mechanical stability would not cause shrinkage or fracture during the freeze-drying process.
Traditional oral contraception needs to be administered for 21 days continuously during a menstrual cycle, which greatly increases the risk of missed medication. The missed medication causes lower progesterone levels in the body, abnormal follicle development, and thin cervical mucus. The rate of unplanned pregnancies due to missed doses is as high as 9% with an increased risk of abnormal fetal growth . Biomaterial-based oral contraception is expected to be self-administration with a lower dosing frequency for up to 1 month. And it will have a lower risk of damage and infection to the gastrointestinal tract due to the superiority of the structural design and antimicrobial properties of biomaterials. For instance, Kirtane et al. proposed that an orally administered long-acting gastric resident dosage form composed of V-shaped six polymer arms loaded LNG and connected by an elastomer core, which allowed it to be folded into a capsule for easy oral administration . After the capsule dissolved in the stomach, the device would form recoils in a larger size than that of the pylorus, so that it could be fixed there and release the drug constantly and steadily (Fig. 6). Importantly, the form consists of Polydimethylsiloxane (PDMS) and poly (sebacic anhydride) to carry LNG for long-term drug release. As an inert polymeric biomaterial, PDMS with good plasticity biocompatibility, biostability, and effective control of drug release has been widely used in a series of drug delivery devices [72,73,74,75]. Poly (anhydride) features not only a larger total drug release but also the release rate that can be adjusted by controlling the monomers used to synthesize the polymer . Pharmacokinetic studies showed that comparable concentrations of LNG were produced on days 3, 11, and 17, revealing a unidirectional variability in serum concentrations. In addition, animal tests showed that 16 of 18 arms remained in the pylorus after 30 days without any detachment. This form is expected to address low drug adherence compared to oral pills and achieve effective contraception for up to 1 month.
The vagina has the unique anatomical advantage of serving as the only route for sperm to enter the woman’s body during intercourse, while the elastic folds of the vagina itself provide the basis for long-term device placement. Compared to the gastrointestinal tract, vaginal long-acting contraception has lower drug interactions and doses of drugs, greatly increasing the biocompatibility and bioavailability of the device. Faster hormone absorption in vaginal epithelium, sustained local release with a lower risk of triggering hormonal side effects made the ring more appealing. In addition, long-acting vaginal contraception can significantly improve user compliance compared to subcutaneous implants, thus ensuring better contraceptive outcomes.
The vaginal ring is an intravaginal cyclic contraceptive device made of hormones together with polymeric compounds with slow release capability . The hormone can be absorbed through the vaginal epithelium into the local or systemic blood circulation, inhibiting follicle development and ovulation, which is safe, reversible without the first-pass effect . It can be placed and removed by the user itself, which greatly improves compliance, but it has the potential to irritate the vagina. Silicone rubber polymer was earlier used as vaginal contraceptive carriers because of the inert silicone elastomer and the larger surface ratio, which allows the carrier to bind more drugs and hormones such as medroxyprogesterone acetate (MPA), LNG, and E2, increasing the duration period of function [79, 80]. Nevertheless, clinical trials showed that silastic rings always presented a high incidence of nausea and vomiting, particularly in the first cycle of use, which attributed to the accumulation of drugs on the surfaces of rings causing the initial burst effect [79, 81]. To address the issue above, polyethylene vinyl acetate copolymers might also update silastic polymer to show into the following technology. Laarhoven et al. presented a vaginal ring consisting of ethylene–vinyl acetate copolymer which exhibited suitable permeability properties that could make a controlled release system with specified release characteristics, values of the diffusion coefficient, and solubility . In this polymer system, it is feasible to load steroids (e.g., ethinylestradiol and ethinylestradiol) to make certain of the effectiveness of the birth control and to reduce the negative effects of each steroid . The application of biodegradable polymers like PLA overcome the responding blockage since the release is controlled by the degradation of the polymer instead of the solubility and subsequent diffusion of the drug through the polymer . Conville et al. demonstrated a hydrogel delivery system consisting of a blend of polyethylene vinyl acetate (PEVA) and polylactic acid with hydrophilic tenofovir, which exhibited a desirable performance in a long-acting slow-release by adjusting the ratio of PLA to PEVA for contraception, pregnancy, and blocking HIV transition . However, hydrogel rings can also present some issues which are always difficult to regulate, such as poor vaginal drug loading and abnormal vaginal discharge . Long et al. proposed a controlled releasing system in which LNG was chemically cross-linked and encapsulated in CS to form CS-LNG microspheres . The microspheres had a concentrated particle size distribution and significantly prolonged the drug release. Besides, the drug release can be more extended and controlled through incorporation into PVA hydrogel with different concentrations (Fig. 7). Saxena et al. fabricated non-biodegradable 2-hydroxyethyl methacrylate and sodium methacrylate, as well as soluble acacia gum hydrogel rings for delivery of anti-HIV agents and non-hormonal contraceptives . The prepared rings showed sustained release for 10 and 28 days in vitro. Nonetheless, a potential risk associated with hydrogel ring is the tendency to absorb aqueous fluids and swell, which can doubtlessly reduce the ability to release drugs and the range of drug interactions with the body in vivo . Sharifzadeh et al. presented a montmorillonite (MMT)-based polyacrylamide hydrogel ring for controlled vaginal drug delivery . MMT sustains drug release via strongly adsorbing to the drug molecule, which improves the bioavailability and dissolution rate of hydrophobic drugs and even more great swelling within controlled in vitro release.
Intrauterine device (IUD)
IUD is a long-acting contraceptive device placed in the uterine cavity, which is classified into copper/hormonal-IUD. It has advantages over other methods in terms of great contraceptive effectiveness, easy administration, low cost, and spontaneous reversibility . The device exerts contraception effect through interfering with fertilized egg implantation, causing a non-inflammatory endometrial response, and preventing binding between sperm and egg. Recently, significant progress has been made in the development of IUDs, with some well-established products like Cu-IUD, hormonal-IUD, frameless IUDs, and unique insertion devices being commercialized (Table 4). However, adverse effects such as uterine bleeding in pregnancy with the device and abnormal vaginal discharge after IUD insertion during IUD use have also been caused widespread concern .
The mechanism of Cu-IUD is considered that the release of Cu2+ causes the inactivation of sperm and the suppression of myometrial contractions . Cu is gradually oxidized and corroded to release Cu2+ after exposure to the uterine cavity . Yet in practice, although the CuT380A is superior to other devices in contraceptive efficacy, bleeding and pain due to drainage and removal were still emphasized . The crucial reason lies in the unstable release rate of Cu2+ and by-products generated in the corrosion process. Direct contact with the uterine fluid causes the rapid oxidation and corrosion of the Cu on the surface of the device. During the initial implantation, Cu2+ is released in amounts of up to 296 μg/day, known as the initial burst release of Cu2+ . When the concentration of Cu2+ exceeds 80 μM in the uterine, it causes spasmodic muscle contractions, which in turn trigger irregular vaginal bleeding, evacuation, and lumbosacral pain . It is necessary for Cu-IUD to ensure a prolonged and effective Cu2+ release concentration. The corrosion of Cu also produces the by-products such as Cu2O, Cu(OH, Cl)2·2(H2O), and Cu2(CO3)(OH)2, which can adhere to the Cu-IUD surface and prevent the release of Cu2+ [100, 101]. The deposits can adhere to the Cu-IUD surface and prevent the release of Cu2+, negatively interfering with the effect of birth control. Therefore, it is imperative to balance the initial burst release of Cu2+ at an early stage with the effective contraceptive concentration over a long period.
A nano-Cu/low-density polyethylene (Nano-Cu/LDPE) as a copper carrier in IUD was reported by Hu et al. . Copper nanoparticles were combined with LDPE by physicochemical methods to form a composite allowing to the nanoparticles be uniformly distributed in the composite material. Its internal spacing provides penetration channels for Cu2+ and corrosion media, and the corrosion rate is effectively controlled by separating the copper nanoparticles from the LDPE, enabling a rapid and constant release rate within 5 h . Besides, tissue toxicity has not been observed in nano-Cu/LDPE during the acute and sub-chronic treatments . It has increased contraceptive effectiveness up to 5 years, which is suitable for the long-acting reversible contraceptive channel . A novel type of alloy made of magnesium (Mg) and ultra-fine grained bulk Cu (UFG-Cu) was also reported to tailor the burst release of Cu2+ in Cu-IUD . Mg exhibits good biocompatibility and degradability and as an essential element of the human body, Mg2+ participate in metabolism and regulate cell proliferation and apoptosis. The addition of Mg to UFG-Cu allows dissolving preferentially compared to Cu, effectively avoiding the explosive release of Cu2+, which can be kept constant during long-term use to ensure effective contraception . In addition, the incorporation of Mg can improve the histocompatibility of Cu-IUD in the uterus with a significant reduction in the side effects of the device. The copolymer (CuCl2/SiO2/PVA) was also synthesized by putting Cu in the form of Cu2+ to overcome the inherent defects of the Cu corrosion process . Polyvinyl alcohol (PVA) is a non-toxic polymer material with great biocompatibility and exhibits minimal cell adhesion and protein absorption ability . Significant improvement in strength, permeability, and biocompatibility after modification by silica sol . In the composite, the chelation equilibrium determines the release of Cu2+, avoiding the formation of oxide by-products and improving the effective utilization of the material . Other biopolymer composites such as chitosan/alginate multilayer film and PDMS-based electropolymeric coating also play effective impacts on controlling the burst-releasing and stability of long-term contraceptive effect, as detailed in Table 5.
Hormonal-IUD is used to prevent pregnancy by containing hormones that interfere with the normal menstrual cycle, and the long-term release of hormones in the uterus suppresses ovulation and changes the properties of the cervical mucus, which is unfavorable to prevent the entry of sperm into the uterus and fertilization. However, the unstable release of hormones has always been a concern. PE has been regarded as a drug delivery matrix for hormonal IUD, which is not only because of its reasonable surface area and release rate after carrying hydrophobic hormones, but also its stiffness and hardness for use in the design of IUDs . PE-progesterone matrix used for IUD as a-helices was be designed by Kalkwarf et al. . However, the progesterone release rate was as high as 225 μg/day/cm2 in the early stage and was relatively constant at 14 μg/day/cm2 until the 6th day . The release rate of 30 μg/day/cm2 in the uterus would prevent contraception without affecting normal ovulation . Therefore, the controlled release of hormones has become a problem to be solved for hormonal-IUD. Nilsson et al. put forward a Norgestrel-PLA film fixed on T-IUD . PLA film is suitable to prepare the long-acting controlled IUD due to the great diffusion and hydrolysis. Clinical trials showed that circulating ovarian function could maintain during intrauterine injection of norgestrel . But the sharp reduction of menstrual blood during the menstrual cycle is one of the main causes it has not been widely used.
The great potential will be explored in the field of personalized contraception through 3D-printing engineering. Factors such as material structure, hormone dosage, and hormone treatment cycles are directly considered in the design to achieve contraceptive purposes better and reduce side effects significantly. A customizable and biodegradable 3D-printed IUD was proposed by Tappa et al. (Fig. 8) . The estrogen and progesterone are encapsulated in PCL, and the intrauterine contraceptive device with a shape that can be adapted to the body's structure is prepared by fused deposition modeling. As a linearly synthesized degradable aliphatic polyester, PCL possesses high mechanical strength, plasticity, biocompatibility, and low degradation rate, while its degradation products are completely metabolized by the tricarboxylic acid cycle or directly excreted by the kidneys, which makes it suitable for long-term carrier implantation in vivo [28, 116]. In addition, PCL has been approved by the U.S. FDA for drug delivery system applications with a good drug delivery structure and stable drug release rate . The 3D-printed IUD showed an extended hormone release over 7 days in vitro . But the steroid concentration of the implants is merely 1% w/w, therefore it will be a meaningful research direction to improve the drug loading capability of 3D-printed IUDs through the superior performance and structural design of biomaterials.
Fallopian tube occlusion
The traditional route of contraception through the fallopian tubes focuses on occluding or removing the tubes permanently, thus preventing the egg from binding to the sperm and the fertilization. The main approach is to inject corrosive drugs into the fallopian tubes, destroying the mucous membrane, permanent scarring, and eventual occlusion of the tube lumen for permanent contraception. However, there is no doubt that side effects, including severe pain, inflammation, tubal perforation, abnormal menstrual cycles, pelvic adhesion, and some reproductive cancer risks, cannot be unavoidable [117, 118]. And there is increasing attention on the use of biomaterials for female tubal occlusion due to its low interference with the female endocrine system, low chance of pelvic infection, and simple and effective operation.
Permanent tubal occlusion
For permanent tubal occlusion, the ideal device must equip properties with high efficiency, great mechanical performance, fabricability, and tissue reactivity . Established tubal occlusion materials, such as silicone rubber, PET, methanoacrylate, and situ-gelling materials formed by Michael-type additive reaction, have been marketed . However, in recent years, researchers have been working on changing the physical state of such materials and combining them with other substances in order to achieve the high contraceptive effect with significantly reduced side effects. Guo et al. proposed that polidocanol were regarded as a detergent-based hardener in the form of foams rather than liquids, which can improve the contact area of the material with the epithelium, increase foam stability while reducing drug concentrations and potentially toxic epithelium-damaging activity . The addition of benzalkonium chloride (BZK) has been shown to improve the stability of PF. Abdala et al. demonstrated a cross-linked type I bovine collagen stent that rapidly absorbed fluid and maintained structural elasticity with uniform pore size, structure, aspect ratio, and distribution for permanent tubal occlusion . Obvious tubal obstruction and peripheral vascularization with degradation of the stent material and luminal filling with immune cells were observed after 12 days . The results showed low cell density and significant collagen deposition in the structure.
Reversible tubal occlusion
Permanent sterilization may cause severe tubal damage and reduced fertility. Therefore, an increasing number of couples are seeking reversible, painless, and safe tubal contraceptive methods. In contrast to silicone rubber alone, Wang et al. proposed a nickel–titanium shape-memory alloy wire as a tubal plug for reversible contraception . The framework consists of silicone rubber and nickel–titanium shape-memory alloy filaments with a cylindrical center and a petal-like Amphi-terminal. The nickel–titanium alloy not only has good biocompatibility, wear resistance, and corrosion resistance, but also allows a combination of perception and shape reversibility [124, 125]. Only 1 of 30 rabbits was pregnant after insertion of the plug, and all animals in the previous contraceptive experiments were pregnant after removal of the plug [123, 126, 127].
In addition to the metallic materials, researchers have attempted to utilize biodegradable polymeric materials as candidates for tubal contraception because of their controlled reversibility. Abdala et al. used ethylene–vinyl alcohol (EVOH) copolymers to plug one of the fallopian tubes in a rabbit model . EVOH copolymers are biocompatible polymers consisting of a random mixture of hydrophobic ethylene and hydrophilic vinyl alcohol. DMSO has diffused away from the mixture when the mixture comes in contact with aqueous media, causing the polymer to settle and solidify in situ and forming a spongy plug that seals the lumen . The copolymer was successfully transferred into the oviducts of all rabbits and no fertilized egg injection sites were observed. However, the only one case was the majority of the embolic agent immediately extruded from the oviduct into the uterus, and histological analysis showed varying degrees of occlusion, fibrosis, and inflammation. Subramanian et al. represented an innovative insight into the use of a hydrogel formulation composed of styrene-cis-butylene-based styrene maleic anhydride (SMA) dissolved in dimethyl sulfoxide (DMSO) to make a non-hormonal tubal contraceptive implant . Polymeric hydrogels cause sperm inactivation by lowering the pH of the surrounding environment to disrupt the acrosome. The low pH not only destroys the acrosome but also alters the morphology of the sperm, leading to low sperm viability and consequently the loss of sperm fertilization. This compound has been widely used for vas deferens injection of reversible male contraceptives and clinical trials have shown a positive trend in birth control [130, 131]. Meanwhile, the hydrogel implanted in the isthmus of the uterus did not show any implantation swelling, dilation, arrest, contraction, or signs of abnormal growth on the outer surface compared to the control group (Fig. 9) .
Vas deferens contraception
Among the traditional male contraceptive methods, condoms, hormonal contraception, and vasectomy have become well-established and effective. However, the estrogenic side effects of steroids and the irreversibility of post-vasectomy have prompted the novel male contraceptive method with characterizations embodying one-time intervention, lengthy-appearing results, low side outcomes, and alternatives of reversibility . Vas occlusion allows for a single intervention, easy administration, good contraceptive effectiveness, and reversible fertility. And the method has almost no side effects on the body and little impact on the sexual experience. More importantly, the choice and design of biomaterials enable the approach to avoid the use of hormones. Light and heat stimulation are alternative surgical options to achieve non-invasive fertility restoration.
Reversible filtering-type intra-vas device (IVD)
In comparison to vasectomy, IVD has both good contraceptive efficiency and the ability to maintain the patency of the vas, which greatly reduces the incidence of complications . Various types of IVD have been developed, such as threads, copper wires, electrical devices, etc. [134,135,136]. Among them, the Cu-IVD shows a good spermicidal effect and antibacterial performance . However, the burst release of copper ions and the formation of CuO on the surface of IVD have been observed, which greatly increases the biological toxicity and limits the long-term use of Cu-IVD. Cross-linked composites containing polyvinyl alcohol (PVA), nanoparticles of silica, and CuCl prepared for IVD can also substantially improve outburst release and metallic copper oxidation. Chen et al. reported that a novel filtering-type IVD was made of nano-SiO2-Cu2+ complex crosslinking polymer composites . Plenty of micropores in the wall of the IVD prototype can be employed to filter the sperm while keeping the vas deferens unobstructed. Besides, the burst release of Cu2+ could be avoided in IVD. The effective utilization of Cu2+ can be achieved in this novel copper-containing composite and it did not cause obvious toxicity for the cells of the male reproductive organs after 1 year of implantation .
Reversible vas occlusion
RISUG® has been reported as a category of male contraception with reversible vas occlusion. SMA copolymer is injected together into the vas deferens, which reduces the pH around the microenvironment and creates a positive charge leading to sperm inactivation [107, 110]. The flakey precipitate is formed in the lumen through injecting RISUG® just like a labyrinth channel when sperm passed brushing under SEM. Additionally, the sperm membrane can be subjected to an electric charge and pH stress brought by the precipitate, impairing acrosome function and causing the leakage of enzymes necessary for fertilization [109, 110]. NaHCO3 and DMSO are both able to dissolve SMA at alkaline pH conditions, which provides a relatively reversible tendency through injecting NaHCO3 and DMSO into vas deferens . Theoretically, DMSO exhibits a highly reactive sulfur portion, which interacts with the ether oxygen (–O–) of SMA to form SMA-DMSO complexes and dimethyl maleic anhydride . Related properties and influences of RISUG® had also been reported in succession (Fig. 10). Measurements in rats showed 100% contraceptive efficiency after 90 days of RISUG® injection and achieved satisfactory reversible fertility after 30 days of NaHCO3 injection [140, 141]. Similar phenomena were also observed in VasalGel™ . The high molecular weight (360 kDa) SMA polymer acts as a physical-charge barrier to block sperm passage. It has been tested in rabbits and rhesus monkeys with satisfactory results in contraception and fertility recovery .
Recently, a new long-lasting contraceptive composite hydrogel system has been able to achieve both physical blockage and chemical inhibition of sperm motility in the vas deferens by sequential injection of calcium alginate hydrogel, PEG-Au nanoparticles (PEG-AuNps), and ethylene diamine tetraacetic acid (EDTA) (Fig. 11) . Sodium alginate, a natural macromolecular material with excellent biocompatibility, can be guided by GDL to form an injectable system with calcium-containing substances . Calcium alginate hydrogel is considered a long-term physical barrier for sperm and retains its good biocompatibility properties. EDTA can not only be used as a calcium chelator in the hydrogel but also inhibit sperm activity . Importantly, PEG-AuNps is regarded as a temperature-increasing agent which can also be solidified at 37 °C to block both ends of EDTA separately. With sequential injections, sodium alginate was cross-linked to form calcium alginate aqueous solids, which acted as a long-term physical barrier, and PEG-AuNps also sealed both ends of the device by curing. When there was a need for fertility recovery, PEG-AuNps was melted after near-infrared light irradiation, allowing EDTA to come into contact with PEG enabling the vas deferens to be unblocked. The contraceptive period can be set directly by adjusting the injection proportion of each reagent in the range of 2–20 weeks. The embolic area can be easily dredged by a short period of non-invasive near-infrared irradiation rather than a second open surgery.
Photothermal nano contraception
The appropriate temperature for sperm formation is slightly lower than the human body temperature. And as the temperature around the testicles increases, sperm production can be disturbed or even stage infertility occurs . Inspired by this theory, an alternative male contraceptive method was developed, which is to inhibit the production of sperm through the photothermal effect of the biomaterial. Li et al. proposed an in situ testicular injection of methoxy poly(ethylene glycol)-modified gold nanorods combined with near-infrared light irradiation to achieve medium to long-term male contraception . As a medical biomaterial approved by the U.S. FDA, gold nanoparticles have been widely used in the fields of drug and gene carriers, diagnostics, imaging, etc. with the superiorities of non-toxicity, good stability, fast metabolism, photothermal effect, antibacterial and antioxidant properties [149,150,151]. In particular, the photothermal effect of gold nanorods has been widely used in the diagnosis and treatment of cancer [152,153,154]. The PEG coating is designed primarily to evade the immune system . It was demonstrated that the heat treatment not exceeding 40 °C with infrared laser irradiation at 808 nm, the morphology of the testes and vas deferens of male rats was partially damaged and the fertility index dropped to ~ 10% on day 7 and recovered to 50% on day 60. In heat treatment at 45 °C, the morphology of the testes and vas deferens was destroyed and the fertility index dropped to 0 on day 7.
However, in situ testicular injection and heat therapy have reduced the patient's acceptance and limited the popularization of this method of contraception. Another potential magnetic hyperthermia treatment with testis targeting and magnetic hyperthermia for male contraception would be described in the next section.
In comparison to other contraceptive routes, intravenous injections have their unique advantages as well as drawbacks. On the one hand, the circulatory system facilitates the diffusion and immune activation of the biomaterial or drug delivery, making it suitable for immuno-contraception, while the intravenous method is well accepted by patients and easy to administer. On the other hand, contraceptive methods that do not aim to activate the immune system require special attention to avoid immune recognition when administered intravenously, and the metabolic pathway and retention time of the material or drug in the body is still not negligible.
In the 1930s, it was reported that the foundation of an active immune state by antigens in human semen might induce female infertility . Immunocontraception has been considered a safe, long-term effective, and reversible method with its main targets of action being sperm–egg interaction and maternal recognition of the fertilized egg . Among them, the main challenge comes from the safety, reliability, and reversibility of active immunity to the reproductive system . Chua et al. demonstrated a chitosan-based nanoplatform for efficient delivery and immune system uptake of luteinizing hormone-releasing hormone (LHRH) . Chitosan-based nanoparticles modified by functional chloroacetyl groups are allowed to covalently attach the peptide-based and protein-based antigen, which was induced high levels of LHRH-specific antibodies. Meanwhile, nanometer-sized particles are more likely to be uptaken by dendritic cells and retained in lymph nodes than micron-sized particles. After 22 days of inoculation of mice with chitosan-based nanoparticles, the levels of LHRH antibodies were similar to those produced after inoculation with antigen-emulsified complete Freund’s adjuvant, demonstrating the contraceptive effectiveness of chitosan-based nanoparticles. D’Souza et al. proposed polyethylene sebacate (PES) encapsulated Sperm-specific 80 kDa Human Sperm Antigen (80 kDa HSA) dissolved in N-methyl-2-pyrrolidone as an alternative to Freund’s adjuvant for peptide vaccines . PES is recognized as a stable degradable polymer with good biocompatibility, non-genotoxic, non-mutagenic properties . And compared with the LHRH antibody, the use of the synthetic peptide HSA not only eliminated the need for antigens from biological sources, but also induced a more specific immune response . The formulation formed PES-based nanoparticles of approximately 100 nm in situ after intradermal injection and achieved an antibody titer up to 1:3200. The loss of sperm motility and head-to-head sperm agglutination was observed in the rabbit, and no severe reaction was detected after treatment. It can be seen that the utilization of nanobiomaterials in immunocontraception significantly improves the safety and efficacy of contraception and reduces the side effects caused by pure immunoadjuvants. However, the irreversibility of immunocontraception cannot be effectively verified, which may make its use cautious in younger age groups.
Elevated serum concentrations of interleukins (e.g. IL-6, IL-1β) can promote embryo implantation in the early phase of pregnancy . Based on this, a down-regulator of interleukin which complexed by nano Piperolactam A (nano PLA)-2-hydroxy-propyl-β-cyclodextrin (HPBCD) was proposed . Compared to natural cyclodextrins, the hydroxyl groups of HPBCD are hydroxypropylated, reducing the strong crystallinity while obtaining a better complexation ability. In addition, the hydrophobic core of HPBCD allows hydrophobic molecules to be trapped in it to achieve sustained drug release . Nano PLA was selected for the down-regulation of interleukin secretion as a new hydrophobic anti-fertility drug of natural origin . Serum levels of IL-6 and IL-1β were not increased in pregnant rats after nano PLA-HPBCD manipulation, while estrogen and progesterone secretion was not affected. In addition, no acute or chronic toxicity was observed during the treatment, indicating the nano PLA-HPBCD was expected to be a nature-derived down-regulator of interleukin for female contraception.
Under normal sexual circumstances, immunocontraception is effective in avoiding pregnancy by preventing gamete production and fertilization by the immune system. However, immunocontraception by suppressing relevant immune targets may increase the probability of autoimmune diseases, which may cause ovarian-related diseases and fail to restore fertility. Nanomaterials are promising for safer, time-controlled contraception through targeted transport of proteins or peptides through the circulatory system to their target sites by embedding, chemical modification, and masking other non-relevant immune targets, avoiding the need to stimulate a systemic immune response .
Magnetic targeting and hyperthermia
As mentioned above, the severe pain from in situ testicular injection and dermal damage caused by near-infrared light irradiation limited the clinical application of photothermal male contraception . Therefore, the nanotechnology of biomaterials for male contraception also requires optimizing the route of operation and avoiding secondary damage. Iron oxide nanoparticles (IONPs) can be used as potential alternative medical nanoparticles due to their biocompatibility, biodegradability, and physiological metabolic pathways [168, 169]. The unique magnetic therapeutic and targeting properties have been widely used in cancer therapy and drug carrier platforms [170,171,172]. Ding et al. proposed controllable male contraception via intravenous administration. PEG@Fe3O4 was synthesized by coating oleic acid through the thermal decomposition method (Fig. 12) . 50 nm of Fe3O4 was selected for exhibiting the best thermal effect on the inhibition of spermatogenesis. After intravenous injection, the nanoparticles were aggregated in the mice testis under the alternating magnetic field while being able to produce a local 37–45 °C environment, effectively reducing the fertility index. Besides, mild magnetic hyperthermia could achieve fertility recovery after 60 days. Compared to direct testicular injection, intravenous injection of nanoparticles would be more acceptable for patients. Importantly, the application of alternating magnetic fields can effectively avoid secondary damage to the skin caused by near-infrared light. Therefore, the combination of targeted delivery with hyperthermia has the potential to enable the clinical translation and application of nanomaterials for contraception and reversible recovery.
Conclusion and outlook
Following the people’s wishes and usage scenarios, an increasing number of long-term reversible contraceptive options and products have been developed. However, the quest for significant contraceptive effectiveness, fewer side effects, and enjoyable sexual experiences during sex still need to be improved. The development of biomaterials for contraceptive technology can not only effectively improve the initial explosive release of drugs and avoid tissue damage and necrosis, but also take advantage of the excellent properties of the materials themselves to achieve effective birth control in a non-hormonal manner. In addition, with the introduction of modified biomaterials and condition-sensitive materials, it is expected that the need for reversible contraception can be further met by a variety of contraceptive methods while ensuring the effectiveness of contraception.
However, to accelerate the translation of biomaterials-based long-acting reversible contraception from the bench to the clinic, more studies are needed to be performed. The development of various types of biomaterials over the past two decades has improved the inferiority of traditional contraceptive methods, with studies showing more effective drug delivery, more stable physical barrier effects, easier reversible recovery, and less surgical damage. However, the need for personalized contraception may be the next research priority. According to the new protocol, some individualized requirements will be able to be significantly enhanced through the introduction of biomaterials with structural optimization in terms of release control, avoidance of hormonal side effects, and conditional degradation. And there may be in short described as follows.
Although there is a significant growth in the use of nanomaterials in contraception through the advantages of drug delivery and controlled drug release, the side effects of nanotoxicity on the human body cannot be ignored. For example, Cu-IUD, which is commonly used in contraception, is absorbed into the bloodstream through local mucosal tissues and induces hepatotoxicity and nephrotoxicity through mitochondrial failure, enhanced ketogenesis, fatty acid beta-oxidation, and glycolysis, resulting in scattered necrosis of hepatocytes and extensive necrosis of the proximal tubules of the kidney . The second factor lies in the small size and surface charge of the nanomaterials, which can pass through the biological barrier, which reduces glutathione levels, inhibits catalase and superoxide dismutase activities to affect reproductive organs adversely . The impacts of the mode of entry of nanomaterials into the body and the structural changes that occur in the body cycle on their biological effects also deserve to be investigated. In addition, nanomaterials can be recognized by the human immune system as foreign antigens causing unpredictable immune attacks and even the departure of autoimmune diseases. In particular, the particle size, surface morphology, and charge of the material play a crucial role in immunotoxicity and may cause reactive oxygen species production, DNA damage, lysosomal damage, mitochondrial dysfunction, and apoptosis or necrosis .
Meanwhile, it is still to be solved that the obstruction can be easily expelled by the peristaltic movement of the muscular duct, which affects the contraceptive effect. Advances in research on structural combinations of biomaterials may improve this deficiency. Structural biomaterial refers to the hierarchical assembly of multiple structural forms of biomaterials for a specific purpose, which makes better use of the advantages among components. As mentioned before, the research about ethylene–vinyl alcohol copolymer and SMA as contraceptive blocker have been made great progress [122, 176]. And on the other hand, scaffolds acted as the support that was constructed from electrospun polycaprolactone and collagen have been shown to perform well in mechanical properties such as tensile strength, maximum load, Young’s modulus, and elongation, which supposed that the tubular scaffold made of biomaterials may be the key point to solve problems . Therefore, a structural biomaterial scaffold might withstand the extrusion of the muscular layer of the pipe preventing the get-away of inner blockages, in addition to protecting the microenvironment within the pipe to attain reversible contraception [178, 179].
Availability of data and materials
World Health Organization
- Cu2 + :
Styrene maleic anhydride
Polylactic acid co-glycolic acid
Mullerian inhibiting substance
Human chorionic gonadotropin
Human chorionic gonadotropin
Depot medroxyprogesterone acetate
Hormonal intrauterine device
Poly (ortho esters)
U.S. Food and Drug Administration
Hydroxypropyl beta cyclodextrin
Polylactic acid co-glycolic acid
Scanning electron microscopy
Polyethylene vinyl acetate
The intrauterine ball
Ultra-fine grained bulk Cu
Ethylene diamine tetraacetic acid
Luteinizing hormone-releasing hormone
Iron oxide nanoparticles
Hirshberg A, Srinivas SK. Epidemiology of maternal morbidity and mortality. Semin Perinatol. 2017;41:332–7.
Fetner T, Dion M, Heath M, Andrejek N, Newell SL, Stick M. Condom use in penile–vaginal intercourse among Canadian adults: results from the sex in Canada survey. PLoS ONE. 2020;15: e0228981.
World Health Organization. Family planning: a global handbook for providers. Geneva: World Health Organization, Johns Hopkins Bloomberg School of Public Health; 2018. p. 559–60.
Thomas S, Visakh P, Mathew AP. Advances in natural polymers. Adv Struct Mater. 2013;18(1):361–96.
Kowalski PS, Bhattacharya C, Afewerki S, Langer RS. Smart biomaterials: recent advances and future directions. ACS Biomater Sci Eng. 2018;4:3809–17.
Laure ISC, Nderson DEA, Lapperich CAMK, Uohung WEK, Ong JOYW. Biomaterials and contraception: promises and pitfalls. Ann Biomed Eng. 2019;48:2113–31.
World Health Organization. Family planning/contraception method. 2020.
Li W, Tang J, Terry RN, Li S, Brunie A, Callahan RL, et al. Long-acting reversible contraception by effervescent microneedle patch. Sci Adv. 2019;5: eaaw8145.
Long J, Etxeberria AE, Kornelsen C, Nand AV, Ray S, Bunt CR, et al. Development of a long-term drug delivery system with levonorgestrel-loaded chitosan microspheres embedded in poly(vinyl alcohol) hydrogel. ACS Appl Bio Mater. 2019;2:2766–79.
Kirtane AR, Hua T, Hayward A, Bajpayee A, Wahane A, Lopes A, et al. A once-a-month oral contraceptive. Sci Transl Med. 2019;11: eaay2602.
Biomaterials. Elsevier; 2020. https://www.journals.elsevier.com/biomaterials. Accessed 5 Jan 2020.
Biomaterials. Nature research. Springer; 2020. https://www.nature.com/subjects/biomaterials. Accessed 5 Jan 2020.
Yang L, Pijuan-Galito S, Rho HS, Vasilevich AS, Eren AD, Ge L, et al. High-throughput methods in the discovery and study of biomaterials and materiobiology. Chem Rev. 2021;121:4561–677.
Li Y, Xiao Y, Liu C. The horizon of materiobiology: a perspective on material-guided cell behaviors and tissue engineering. Chem Rev. 2017;117:4376–421.
Yang B, Chen Y, Shi J. Reactive oxygen species (ROS)-based nanomedicine. Chem Rev. 2019;119:4881–985.
Zhu Y, Liu L, Sun Z, Ji Y, Wang D, Mei L, et al. Fucoidan as a marine-origin prebiotic modulates the growth and antibacterial ability of Lactobacillus rhamnosus. Int J Biol Macromol. 2021;180:599–607.
Hao Y, Zheng W, Sun Z, Zhang D, Sui K, Shen P, et al. Marine polysaccharide-based composite hydrogels containing fucoidan: preparation, physicochemical characterization, and biocompatible evaluation. Int J Biol Macromol. 2021;183:1978–86.
Yin X, Hao Y, Lu Y, Zhang D, Zhao Y, Mei L, et al. Bio-multifunctional hydrogel patches for repairing full-thickness abdominal wall defect. Adv Funct Mater. 2021;31:2105614.
Zhou Q, Chen J, Luan Y, Vainikka PA, Thallmair S, Marrink SJ, et al. Unidirectional rotating molecular motors dynamically interact with adsorbed proteins to direct the fate of mesenchymal stem cells. Sci Adv. 2020;6: eaay2756.
He Y, Zhao W, Dong Z, Ji Y, Li M, Hao Y, et al. A biodegradable antibacterial alginate/carboxymethyl chitosan/Kangfuxin sponges for promoting blood coagulation and full-thickness wound healing. Int J Biol Macromol. 2021;167:182–92.
Liu L, Han Z, An F, Gong X, Zhao C, Zheng W, et al. Aptamer-based biosensors for the diagnosis of sepsis. J Nanobiotechnology. 2021;19:216.
Zheng W, Hao Y, Wang D, Huang H, Guo F, Sun Z, et al. Preparation of triamcinolone acetonide-loaded chitosan/fucoidan hydrogel and its potential application as an oral mucosa patch. Carbohydr Polym. 2021;272: 118493.
Ding W, Chen Z, Gu Y, Chen Z, Zheng Y, Sun F. Magnetic testis targeting and magnetic hyperthermia for noninvasive, controllable male contraception via intravenous administration. Nano Lett. 2021;21:6289–97.
Brahmi D, Steenland MW, Renner RM, Gaffield ME, Curtis KM. Pregnancy outcomes with an IUD in situ: a systematic review. Contraception. 2012;85:131–9.
Croxatto HB. Progestin implants for female contraception. Contraception. 2002;65(1):15–9.
Kalkwarf DR, Sikov MR, Smith L, Gordon R. Release of progesterone from polyethylene devices in vitro and in experimental animals. Contraception. 1972;6:423–31.
Carey RF, Herman WA, Retta SM, Rinaldi JE, Herman BA, Athey TW. Effectiveness of latex condoms as a barrier to human immunodeficiency virus-sized particles under conditions of simulated use. Sex Transm Dis. 1992;19:230–4.
Subramanian B, Agarwal T, Roy A, Parida S, Kundu B, Maiti TK, et al. Synthesis and characterization of PCL-DA:PEG-DA based polymeric blends grafted with SMA hydrogel as bio-degradable intrauterine contraceptive implant. Mater Sci Eng C. 2020;116: 111159.
Bao W, Xie L, Zeng X, Kang H, Wen S, Cui B, et al. A cocktail-inspired male birth control strategy with physical/chemical dual contraceptive effects and remote self-cleared properties. ACS Nano. 2019;13:1003–11.
Yao G, Quan G, Lin S, Peng T, Wang Q, Ran H, et al. Novel dissolving microneedles for enhanced transdermal delivery of levonorgestrel: in vitro and in vivo characterization. Int J Pharm. 2017;534:378–86.
Hess RA, Bunick D, Lee K, Bahr J, Taylor JA, Kenneth S, et al. A role for oestrogens in the male reproductive system. Nature. 1997;390:509–12.
Kavanaugh ML, Jerman J. Contraceptive method use in the United States: trends and characteristics between 2008, 2012 and 2014. Contraception. 2018;97:14–21.
Creasy DM, Chapin RE. Male reproductive system. In: Haschek and Rousseaux’s handbook of toxicologic pathology. Amsterdam: Elsevier; 2013. p. 2493–598.
De KDM, Loveland KL, Meinhardt A, Simorangkir D, Wreford N. Spermatogenesis. Hum Reprod. 1998;13:1–8.
Vogazianou A. Anatomy and physiology of the female reproductive system. In: Advanced practice in endocrinology nursing. Cham: Springer; 2019. p. 739–52.
Williamson LM, Parkes A, Wight D, Petticrew M, Hart GJ. Limits to modern contraceptive use among young women in developing countries: a systematic review of qualitative research. Reprod Health. 2009;6:1–12.
Aranha C, Gupta S, Reddy KVR. Contraceptive efficacy of antimicrobial peptide Nisin: in vitro and in vivo studies. Contraception. 2004;69:333–8.
Hapgood JP, Kaushic C, Hel Z. Hormonal contraception and HIV-1 acquisition: biological mechanisms. Endocr Rev. 2018;39:36–78.
Urrutia RP, Polis CB, Jensen ET, Greene ME, Kennedy E, Stanford JB. Effectiveness of fertility awareness-based methods for pregnancy prevention a systematic review. Obstet Gynecol. 2018;132:591–604.
Cleland J, Conde-agudelo A, Peterson H, Ross J, Tsui A. Contraception and health. Lancet. 2012;380:149–56.
Mansour D, Inki P, Gemzell-Danielsson K. Efficacy of contraceptive methods: a review of the literature. Eur J Contracept Reprod Healthc. 2010;15:4–16.
Graham JD, Clarke CL. Physiological action of progesterone in target tissues. Endocr Rev. 1997;18:502–19.
Grandi G, Farulla A, Sileo FG, Facchinetti F. Levonorgestrel-releasing intra-uterine systems as female contraceptives. Expert Opin Pharmacother. 2018;19:677–86.
Ramdhan RC, Simonds E, Wilson C, Loukas M, Oskouian RJ, Tubbs RS. Complications of subcutaneous contraception: a review. Cureus. 2018;10: e2132.
Fuchs R, Taylor D, Jenkins DW, Brache V, Luo D, Dor LJ, et al. Levonorgestrel release rates measured through analysis of two-rod contraceptive explants. Contracept X. 2020;2: 100039.
Palomba S, Falbo A, Di Cello A, Materazzo C, Zullo F. Nexplanon: the new implant for long-term contraception. A comprehensive descriptive review. Gynecol Endocrinol. 2012;28:710–21.
Widder KJ, Green R. Drug and enzyme targeting. Orlando: Academic Press; 1985.
Heller J. Controlled drug release from poly (ortho esters)—a surface eroding polymer. J Control Release. 1985;2:167–77.
Zignani M, Merkli A, Sintzel MB, Bernatchez SF, Kloeti W, Heller J, et al. New generation of poly (ortho esters): synthesis, characterization, kinetics, sterilization and biocompatibility. J Control Release. 1997;48:115–29.
Getme AS, Patel B. A review: bio-fiber’s as reinforcement in composites of polylactic acid (PLA). Mater Today Proc. 2020;26:2116–22.
Elsawy MA, Kim K, Park J, Deep A. Hydrolytic degradation of polylactic acid ( PLA ) and its composites. Renew Sustain Energy Rev. 2017;79:1346–52.
Athanasiou KA, Niederauer GG, Agrawal CM. Sterilization, toxicity, biocompatibility and clinical applications of polylactic acid/ polyglycolic acid copolymers. Biomaterials. 1996;17:93–102.
Gavasane AJ, Pawar HA. Synthetic biodegradable polymers used in controlled drug delivery system: clinical pharmacology synthetic biodegradable polymers used in controlled drug delivery system: an overview. Clin Pharmacol Biopharm. 2014;3:1000121.
Sun H, Mei L, Song C, Cui X, Wang P. The in vivo degradation, absorption and excretion of PCL-based implant. Biomaterials. 2006;27:1735–40.
Bahl S, Dolma J, Jyot J, Sehgal S. Proceedings biodegradation of plastics: a state of the art review. Mater Today Proc. 2021;39:31–4.
Benagiano G, Gabelnick H, Farris M. Contraceptive devices: subcutaneous delivery systems. Expert Rev Med Dev. 2008;5:623–37.
Kolawole OO, Sowemimo OO, Ojo OO, Fasubaa OB. Contraceptive implants: a review and current perspective in southwest Nigeria. Trop J Obstet Gynaecol. 2018;35:108–12.
Patel D, Chaudhary SA, Parmar B, Bhura N. Transdermal drug delivery system: a review. J Pharm Res. 2012;1:66–75.
Zieman M, Guillebaud J, Weisberg E, Shangold GA, Fisher AC, Creasy GW. Contraceptive efficacy and cycle control with the Ortho Evra™/Evra™ transdermal system: the analysis of pooled data. Fertil Steril. 2002;77:S13–8.
Archer DF, Cullins V, Creasy GW, Fisher AC. The impact of improved compliance with a weekly contraceptive transdermal system (Ortho Evra®) on contraceptive efficacy. Contraception. 2004;69:189–95.
Ingrole RSJ, Azizoglu E, Dul M, Birchall JC, Gill HS, Prausnitz MR. Trends of microneedle technology in the scientific literature, patents, clinical trials and internet activity. Biomaterials. 2020;267: 120491.
Ma G, Wu C. Microneedle, bio-microneedle and bio-inspired microneedle: a review. J Control Release. 2017;251:11–23.
Saokham P, Muankaew C, Jansook P. Solubility of cyclodextrins and drug/cyclodextrin complexes. Molecules. 2018;23:1161.
Yao AG, Quan G, Lin S. Novel dissolving microneedles for enhanced transdermal delivery of levonorgestrel: in vitro and in vivo characterization. Int J Pharm. 2017;534:378–86.
Elmowafy EM, Tiboni M, Soliman ME. Biocompatibility, biodegradation and biomedical applications of poly (lactic acid)/poly (lactic-co-glycolic acid ) micro and nanoparticles. J Pharm Investig. 2019;49:347–80.
Wang C, Jiang X, Zeng Y, Terry RN, Li W. Rapidly separable microneedle patches for controlled release of therapeutics for long-acting therapies. Med Drug Discov. 2021;13: 100118.
Bao L, Park J, Bonfante G, Kim B. Recent advances in porous microneedles: materials, fabrication, and transdermal applications. Drug Deliv Transl Res. 2022;12:395–414.
Schwingl PJ, Guess HA. Safety and effectiveness of vasectomy. Fertil Steril. 2000;73:923–36.
Murdoch FE, Goldberg E. Male contraception: another holy grail. Bioorganic Med Chem Lett. 2014;24:419–24.
Feng Y, Wang K, Yao J, Webley PA, Smart S, Wang H. Effect of the addition of polyvinylpyrrolidone as a pore-former on microstructure and mechanical strength of porous alumina ceramics. Ceram Int. 2013;39:7551–6.
Zapata LB, Steenland MW, Brahmi D, Marchbanks PA, Curtis KM. Effect of missed combined hormonal contraceptives on contraceptive effectiveness: a systematic review. Contraception. 2013;87:685–700.
Tang Q, Li X, Lai C, Li L, Wu H, Wang Y, et al. Fabrication of a hydroxyapatite-PDMS microfluidic chip for bone-related cell culture and drug screening. Bioact Mater. 2021;6:169–78.
Holländer J, Hakala R, Suominen J, Moritz N, Yliruusi J. 3D printed UV light cured polydimethylsiloxane devices for drug delivery. Int J Pharm. 2018;544:433–42.
Zhao X, Liu Y, Shao C, Nie M, Huang Q, Li J. Photoresponsive delivery microcarriers for tissue defects repair. Adv Sci. 2019;6: 190280.
Kumar A, Singh P, Rai B, Sharma A, Sivakumar S. Stable sub-100 nm PDMS nanoparticles as an intracellular drug delivery vehicle. Mater Sci Eng C. 2021;119: 111577.
Kirtane AR, Abouzid O, Minahan D, Bensel T, Hill AL, Selinger C, et al. Development of an oral once-weekly drug delivery system for HIV antiretroviral therapy. Nat Commun. 2018;9:1–12.
Kerns J, Darney P. Vaginal ring contraception. Contraception. 2011;83:107–15.
Wilhelmus M, Den HV, Jacoba A, Van BM, Kafi A, Alnabawy M, et al. Comparison of ethinylestradiol pharmacokinetics in three hormonal contraceptive formulations: the vaginal ring, the transdermal patch and an oral contraceptive. Contraception. 2005;72:168–74.
Roy S, Krauss RM, Mishell DR, Casagrande J, Pike MC. The effect on lipids and lipoproteins of a contraceptive vaginal ring containing levonorgestrel and estradiol. Contraception. 1981;24:429–49.
Mishell DR, Lumkin ME. Contraceptive effect of varying dosages of progestogen in silastic vaginal rings. Fertil Steril. 1970;21:99–103.
Weisberg E, Fraser IS, Lacarra M, Mishell DR, Jackanicz T. Effect of different insertion regimens on side effects with a combination contraceptive vaginal ring. Contraception. 1997;56:233–9.
Van Laarhoven JAH, Kruft MAB, Vromans H. In vitro release properties of etonogestrel and ethinyl estradiol from a contraceptive vaginal ring. Int J Pharm. 2002;232:163–73.
Nguyen BT, Jensen JT. Evaluating the efficacy and safety of a progestin- and estrogen-releasing ethylene vinyl acetate copolymer contraceptive vaginal ring. Expert Opin Drug Saf. 2014;13:1423–30.
Kenawy ER, Abdel-Hay FI, El-Newehy MH, Wnek GE. Processing of polymer nanofibers through electrospinning as drug delivery systems. Mater Chem Phys. 2009;113:296–302.
Conville CM, Major I, Friend DR, Clark MR, Woolfson AD, Malcolm RK. Development of polylactide and polyethylene vinyl acetate blends for the manufacture of vaginal rings. J Biomed Mater Res B Appl Biomater. 2012;100:891–5.
Hussain A, Ahsan F. The vagina as a route for systemic drug delivery. J Control Release. 2005;103:301–13.
Han YA, Singh M, Saxena BB. Development of vaginal rings for sustained release of nonhormonal contraceptives and anti-HIV agents. Contraception. 2007;76:132–8.
Malcolm RK, Boyd PJ, McCoy CF, Murphy DJ. Microbicide vaginal rings: technological challenges and clinical development. Adv Drug Deliv Rev. 2016;103:33–56.
Sharifzadeh G, Hezaveh H, Muhamad II, Hashim S, Khairuddin N. Montmorillonite-based polyacrylamide hydrogel rings for controlled vaginal drug delivery. Mater Sci Eng C. 2020;110: 110609.
Grimes DA. Intrauterine device and upper-genital-tract infection. Lancet. 2000;356:1013–9.
Patchen L, Berggren EK. Use of the copper T380A intrauterine device by adolescent mothers: continuation and method failure. J Pediatr Adolesc Gynecol. 2011;24:71–3.
Ti AJ, Roe AH, Whitehouse KC, Smith RA, Gaffield ME, Curtis KM. Effectiveness and safety of extending intrauterine device duration: a systematic review. Am J Obstet Gynecol. 2020;223:24–35.
Nelson AL, Massoudi N. New developments in intrauterine device use: focus on the US. Open Access J Contracept. 2016;7:127–41.
Kapp N, Curtis KM. Intrauterine device insertion during the postpartum period: a systematic review. Contraception. 2009;80:327–36.
Oster GK. Chemical reactions of the copper intrauterine device. Fertil Steril. 1972;23:18–23.
Bastidas DM, Valdez B, Schorr M, Bastidas JM. Corrosion of copper intrauterine devices: review and recent developments. Corros Rev. 2019;37:307–20.
Kulier R, Helmerhorst F, O’Brien P, Usher-Patel M, D’Arcangues C. Copper containing intra-uterine devices for contraception. Cochrane Database Syst Rev. 2005;3:CD005347.
Xue H, Xu N, Zhang C. Corrosion behavior of copper in a copper bearing intrauterine device in the presence of indomethacin. Contraception. 1998;57:49–53.
Sanders JN, Adkins DE, Kaur S, Storck K, Gawron M, Turok DK. Bleeding, cramping, and satisfaction among new copper IUD users: a prospective study. PLoS ONE. 2018;13: e0199724.
Xu XX, Nie FL, Wang YB, Zhang JX, Zheng W, Li L, et al. Effective inhibition of the early copper ion burst release with ultra-fine grained copper and single crystal copper for intrauterine device application. Acta Biomater. 2012;8:886–96.
Wang Z, Wang X, Wang Y, Zhu Y, Liu X, Zhou Q. NanoZnO-modified titanium implants for enhanced anti-bacterial activity, osteogenesis and corrosion resistance. J Nanobiotechnol. 2021;19:353.
Lei H, Duan Z. Studies of acute and subchronic systemic toxicity associated with a copper/low-density polyethylene nanocomposite intrauterine device. Int J Nanomed. 2018;13:4913–26.
Luyt AS, Molefi JA, Krump H. Thermal, mechanical and electrical properties of copper powder filled low-density and linear low-density polyethylene composites. Polym Degrad Stab. 2006;91:1629–36.
Xia X, Tang Y, Xie C, Wang Y, Cai S, Zhu C. An approach to give prospective life-span of the copper/low-density-polyethylene nanocomposite intrauterine device. J Mater Sci Mater Med. 2011;22:1773–81.
Fan Q, Bao G, Ge D, Wang K, Sun M, Liu T, et al. Effective easing of the side effects of copper intrauterine devices using. Acta Biomater. 2021;128:523–39.
Li J, Suo J, Huang X, Ye C, Wu X. Release behavior of copper ion in a novel contraceptive composite. Contraception. 2007;76:233–7.
Burczak K, Gamian E, Kochman A. Long-term in tivo performance and biocompatibility of poly(viny1 alcohol) hydrogel macrocapsules for hybrid-type artificial pancreas. Biamateriols. 1996;17:2351–6.
Li J, Suo J, Zou P, Jia L, Wang S. Structure, corrosion behavior and mechanical property of a novel poly (vinyl alcohol) composite in simulated body fluid. J Biomater Sci. 2010;21:863–76.
Tian K, Xie C, Xia X. Biointerfaces chitosan/alginate multilayer film for controlled release of IDM on Cu/LDPE composite intrauterine devices. Colloids Surf B Biointerfaces. 2013;109:82–9.
Xu XX, Ding MH, Zhang JX, Zheng W, Li L, Zheng YF. A novel copper/polydimethiylsiloxane nanocomposite for copper-containing intrauterine contraceptive devices. J Biomed Mater Res B Appl Biomater. 2014;101:1428–36.
Ramakrishnan R, Bharaniraja B, Aprem AS. Controlled release of copper from an intrauterine device using a biodegradable polymer. Contraception. 2015;92:585–8.
Bertuola M, Grillo CA, Fernández Lorenzo de Mele M. Eradication of burst release of copper ions from copper-bearing IUDs by a phytocompound-based electropolymeric coating. Mater Lett. 2019;252:317–20.
Ronca S. Polyethylene. In: Brydson’s plastics materials. Amsterdam: Elsevier; 2017. p. 247–78.
Rice CF, Killick SR, Dieben T, Bennink HC. A comparison of the inhibition of ovulation achieved by desogestrel 75 µg and levonorgestrel 30 µg daily. Hum Reprod. 1999;14:982–5.
Tappa K, Jammalamadaka U, Ballard DH, Bruno T, Israel MR, Vemula H, et al. Medication eluting devices for the field of OBGYN (MEDOBGYN): 3D printed biodegradable hormone eluting constructs, a proof of concept study. PLoS ONE. 2017;12: e0182929.
Mkhabela VJ, Ray SS. Poly(ε-caprolactone) nanocomposite scaffolds for tissue engineering: a brief overview. J Nanosci Nanotechnol. 2014;14:535–45.
Kjaer SK, Mellemkjaer L, Brinton LA, Johansen C, Gridley G, Olsen JH. Tubal sterilization and risk of ovarian, endometrial and cervical cancer. A Danish population-based follow-up study of more than 65 000 sterilized women. Int J Epidemiol. 2004;33:596–602.
Erruo W, Bilian X, Weiqian Y, Hui L, Beisheng W. Hormonal profile of the menstrual cycle in Chinese women after tubal sterilization. Contraception. 1992;45:583–93.
Mcswain H, Brodie MF. Fallopian tube occlusion, an alternative to tubal ligation. Tech Vasc Interv Radiol. 2006;9:24–9.
Gormley R, Vickers B, Cheng B, Norman WV. Comparing options for females seeking permanent contraception in high resource countries: a systematic review. Reprod Health. 2021;18:154.
Guo JX, Lucchesi L, Gregory KW. Improvement of stability of polidocanol foam for nonsurgical permanent contraception. Contraception. 2015;92:103–7.
Abdala N, Levitin A, Dawson A, Maffra R, Munoz-Ramirez H, Godec K, et al. Use of ethylene vinyl alcohol copolymer for tubal sterilization by selective catheterization in rabbits. J Vasc Interv Radiol. 2001;12:979–84.
Wang LG, Qiu Y, Fan YJ, Li XY, Han XJ. Reversible contraceptive effect of the oviduct plug with nickel–titanium shape memory alloy and silicone rubber in rabbits. Contraception. 2011;83:373–7.
Wever DJ, Veldhuizen AG, Sanders MM, Schakenraad JM, Van Horn JR. Cytotoxic, allergic and genotoxic activity of a nickel–titanium alloy. Biomaterials. 1997;18:1115–20.
Ryhänen J, Niemi E, Serlo W, Niemelä E, Sandvik P, Pernu H, et al. Biocompatibility of nickel–titanium shape memory metal and its corrosion behavior in human cell cultures. J Biomed Mater Res. 1997;35:451–7.
Zhao X, Zhu Z, Sun Y, Wang Y. Nickel–titanium shape-memory sawtooth-arm embracing fixator for periprosthetic femoral fractures. Int Orthop. 2012;36:619–26.
Qiu Y, Wang L, Zhang M, Zhang Y, Zhang A, Yang D. A new experimental three-dimensional, reticular intrauterine device (3-DRIUD) composed of nitinol and silicone rubber. Contraception. 2013;88:31–6.
Maubon J, Scanlan M, Thurmond S, Rouanet P. Selective tubal sterilization in rabbits: experience with a hrdrogel combined with a sclerosing agent. Radiology. 1994;193:721–3.
Subramanian B, Agarwal T, Roy A, Parida S. Synthesis and characterization of PCL-DA: PEG-DA based polymeric blends grafted with SMA hydrogel as bio-degradable intrauterine contraceptive implant. Mater Sci Eng C. 2020;116: 111159.
Guha SK. Biophysical mechanism-mediated time-dependent effect on sperm of human and monkey vas implanted polyelectrolyte contraceptive. Asian J Androl. 2007;9:221–7.
Sharma U, Chaudhury K, Jagannathan NR, Guha SK. A proton NMR study of the effect of a new intravasal injectablemale contraceptive RISUG on seminal plasma metabolites. Reproduction. 2001;122:431–6.
Long JE, Lee MS, Blithe DL. Male contraceptive development: update on novel hormonal and nonhormonal methods. Clin Chem. 2019;65:153–60.
Chen Z, Huang X, Suo J, Li J, Sun L. The contraceptive effect of a novel filtering-type nano-copper complex/polymer composites intra-vas device on male animals. Int J Endrol. 2010;33:810–7.
Misro MM, Kaur H, Mahajan S, Guha SK. An intravasal non-occlusive contraceptive device in rats. J Reprod Fertil. 1982;65:9–13.
Song L, Gu Y, Lu W, Liang X, Chen Z. A phase II randomized controlled trial of a novel male contraception, an intra-vas device. Int J Androl. 2006;29:489–95.
Huang X, Suo J, Chen C, Du Q, Shen J, Zhou J. Initial studies on a novel filtering-type intra-vas device in male dogs. Contraception. 2010;81:350–4.
Intravas C, Ivd D, Contraception M. Copper intravas device (IVD) and male contraception. Contraception. 1984;29:45–54.
Chaube SK, Misro MM. Effect of styrene maleic anhydride (SMA) on viability and integrity of isolated rat oocytes in vitro. Contraception. 2002;66:469–72.
Anhydride M, Koul V, Srivastav A, Guha SK. Reversibility with sodium bicarbonate. Contraception. 1998;7824:227–31.
Gordon DKSS. Release of progesterone from polyethylene devices in vitro and in experimental animals. Contraception. 1972;6:423–31.
Luukkainen CNJJ. Biodegradable polylactate as a steroid-releasing polymer: intrauterine administration of d-norgestrel. Am J Obstet Gynecol. 1975;122:90–5.
Thirumalai A, Page ST. Recent developments in male contraception. Drugs. 2019;79:11–20.
Wallee D, Bolick D, Lissner E, Premanandan C, Gamerman G. Reversibility of Vasalgel™ male contraceptive in a rabbit model. Basic Clin Androl. 2017;27:8.
Kumar S, Roy S, Chaudhury K, Sen P, Guha SK. Study of the micro-structural properties of RISUG®—a newly developed male contraceptive. J Biomed Mater Res B Appl Biomater. 2008;86:154–61.
Sharma S, Sen P, Mukhopadhyay SN, Guha SK. Microbicidal male contraceptive—Risug induced morphostructural damage in E. coli. Colloids Surf B Biointerfaces. 2003;32:43–50.
Roy S, Ghosh D, Guha SK. Polyelectrolyte polymer properties in relation to male contraceptive RISUG® action. Colloids Surf B Biointerfaces. 2009;69:77–84.
Durairajanayagam D, Agarwal A, Ong C. Causes, effects and molecular mechanisms of testicular heat stress. Reprod Biomed Online. 2015;30:14–27.
Li WQ, Sun CY, Wang F, Wang YC, Zhai YW, Liang M, et al. Achieving a new controllable male contraception by the photothermal effect of gold nanorods. Nano Lett. 2013;13:2477–84.
Pissuwan D, Niidome T. Gold nanoparticles for the development of transdermal delivery systems. ACS Symp Ser. 2012. p. 69–80.
Jazmín Silvero MC, Rocca DM, Artur de la Villarmois E, Fournier K, Lanterna AE, Perez MF, et al. Selective photoinduced antibacterial activity of amoxicillin-coated gold nanoparticles: from one-step synthesis to in vivo cytocompatibility. ACS Omega. 2018;3:1220–30.
Hu X, Zhang Y, Ding T, Liu J, Zhao H. Multifunctional gold nanoparticles: a novel nanomaterial for various medical applications and biological activities. Front Bioeng Biotechnol. 2020;8:1–17.
Choi WI, Kim JY, Kang C, Byeon CC, Kim YH, Tae G. Tumor regression in vivo by photothermal therapy based on gold-nanorod-loaded, functional nanocarriers. ACS Nano. 2011;5:1995–2003.
Kuo W-S, Chang C-N, Chang Y-T, Yang M-H, Chien Y-H, Chen S-J, et al. Gold nanorods in photodynamic therapy, as hyperthermia agents, and in near-infrared optical imaging. Angew Chem. 2010;122:2771–5.
Choi WI, Sahu A, Kim YH, Tae G. Photothermal cancer therapy and imaging based on gold nanorods. Ann Biomed Eng. 2012;40:534–46.
Ball P. Nano contraception. Nat Mater. 2013;12:602.
Baskin MJ. Temporary sterilization by the injection of human spermatozoa. A preliminary report. Am J Obstet Gynecol. 1932;24:892–7.
Aitken RJ. Immunocontraceptive vaccines for human use. J Reprod Immunol. 2002;57:273–87.
Chu BY, Al Kobiasi M, Zeng W, Mainwaring D, Jackson DC. Chitosan-based particles as biocompatible delivery vehicles for peptide and protein-based vaccines. Procedia Vaccinol. 2012;6:74–9.
D’Souza AA, Yevate SV, Bandivdekar AH, Devarajan PV. In situ polyethylene sebacate particulate carriers as an alternative to Freund’s adjuvant for delivery of a contraceptive peptide vaccine—a feasibility study. Int J Pharm. 2015;496:601–8.
Guhagarkar SA, Malshe VC, Devarajan PV. Nanoparticles of polyethylene sebacate: a new biodegradable polymer. AAPS PharmSciTech. 2009;10:935–42.
Azmi F, Fuaad AAHA, Skwarczynski M, Toth I. Recent progress in adjuvant discovery for peptide-based subunit vaccines. Hum Vaccines Immunother. 2014;10:778–96.
Lu H, Wu JY, Kudo T, Ohno T, Graham DY, Yamaoka Y, et al. Regulation of interleukin-6 promoter activation in gastric epithelial cells infected with Helicobacter pylori. Mol Biol Cell. 2005;16:4954–66.
Bhattacharya P, Saha A, Basak S. Discovery of nano-piperolactam A: a nonsteroidal contraceptive lead acting through down-regulation of interleukins. Nanomed Nanotechnol Biol Med. 2019;18:347–58.
Yokoo M, Kubota Y, Motoyama K, Higashi T. 2-Hydroxypropyl-β-cyclodextrin acts as a novel anticancer agent. PLoS ONE. 2015;10: e0141946.
Sharma JD, Sharma L, Yadav P. Antifertility efficacy of Piper betle Linn. (Petiole) on female albino rats. Asian J Exp Sci. 2017;21:145–50.
Yu Z, Li Q, Wang J, Yu Y, Wang Y, Zhou Q, et al. Reactive oxygen species-related nanoparticle toxicity in the biomedical field. Nanoscale Res Lett. 2020;15:115.
Chen Y, Jiang L, Wang R, Lu M, Zhang Q, Zhou Y, et al. Injectable smart phase-transformation implants for highly efficient in vivo magnetic-hyperthermia regression of tumors. Adv Mater. 2014;26:7468–73.
Espinosa A, Di Corato R, Kolosnjaj-Tabi J, Flaud P, Pellegrino T, Wilhelm C. Duality of iron oxide nanoparticles in cancer therapy: amplification of heating efficiency by magnetic hyperthermia and photothermal bimodal treatment. ACS Nano. 2016;10:2436–46.
Larsen EKU, Nielsen T, Wittenborn T, Rydtoft LM, Lokanathan AR, Hansen L, et al. Accumulation of magnetic iron oxide nanoparticles coated with variably sized polyethylene glycol in murine tumors. Nanoscale. 2012;4:2352–61.
Alphandéry E. Bio-synthesized iron oxide nanoparticles for cancer treatment. Int J Pharm. 2020;586: 119472.
Alphandéry E. Light-interacting iron-based nanomaterials for localized cancer detection and treatment. Acta Biomater. 2021;124:50–71.
Alphandéry E. Iron oxide nanoparticles for therapeutic applications. Drug Discov Today. 2020;25:141–9.
Marmiroli M, Pagano L, Rossi R, De La Torre-Roche R, Lepore GO, Ruotolo R, et al. Copper oxide nanomaterial fate in plant tissue: nanoscale impacts on reproductive tissues. Environ Sci Technol. 2021;55:10769–83.
Roychoudhury S, Nath S, Massanyi P, Stawarz R, Kacaniova M, Kolesarova A. Copper-induced changes in reproductive functions: in vivo and in vitro effects. Physiol Res. 2016;65:11–22.
Hussain Z, Thu HE, Haider M, Khan S, Sohail M, Hussain F, et al. A review of imperative concerns against clinical translation of nanomaterials: unwanted biological interactions of nanomaterials cause serious nanotoxicity. J Drug Deliv Sci Technol. 2020;59: 101867.
Subramanian B, Rameshbabu AP, Ghosh K, Jha PK, Jha R, Murugesan S, et al. Impact of styrene maleic anhydride (SMA) based hydrogel on rat fallopian tube as contraceptive implant with selective antimicrobial property. Mater Sci Eng C. 2019;94:94–107.
Ju YM, Choi JS, Atala A, Yoo JJ, Lee SJ. Bilayered scaffold for engineering cellularized blood vessels. Biomaterials. 2010;31:4313–21.
Wu Y, Wang L, Zhao X, Hou S, Guo B, Ma PX. Self-healing supramolecular bioelastomers with shape memory property as a multifunctional platform for biomedical applications via modular assembly. Biomaterials. 2016;104:18–31.
Fomby P, Cherlin AJ, Hadjizadeh A, Doillon CJ, Sueblinvong V, Weiss DJ, et al. Stem cells and cell therapies in lung biology and diseases: conference report. Ann Am Thorac Soc. 2010;12:181–204.
The authors are very grateful for the financial support of the National Natural Science Foundation of China (Grant No. 31900957), Shandong Provincial Natural Science Foundation (Grant No. ZR2019QC007), Innovation and technology program for the excellent youth scholars of higher education of Shandong province (Grant No. 2019KJE015), and Traditional Chinese Medicine Science and Technology Project of Shandong province (Grant No. 2021Q069).
Ethics approval and consent to participate
Consent for publication
All authors consent to publish the manuscript in Journal of Nanobiotechnology.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Yan, M., Zhang, Y., Wu, Z. et al. Recent progress in advanced biomaterials for long-acting reversible contraception. J Nanobiotechnol 20, 138 (2022). https://doi.org/10.1186/s12951-022-01329-5
- Long-acting reversible contraception
- Biodegradable implant
- Delivery route
- Controlled drug delivery